-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu May 18 22:19:17 2023 UTC
-- File 20230518_102637_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.785955 to 91.055413
-each separately from 7s to 91s and from 55124us to 971010us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.745772 in 1 packets: 18
      8.197660 in 1 packets: 36
      21.234412 in 1 packets: 54
      15.005766 in 1 packets: 72
      7.247931 in 1 packets: 90
      14.835894 in 1 packets: 108
MIU packet time from 2023-05-18T10:26:47.785.955.000 to 2023-05-18T10:28:11.055.413.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.745772 in 1 packets: 18
      8.197660 in 1 packets: 36
      21.234412 in 1 packets: 54
      15.005766 in 1 packets: 72
      7.247931 in 1 packets: 90
      14.835894 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.7615827 to 92.7612770
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999381 in 1 packets: 18
      7.9999710 in 1 packets: 36
      20.9999234 in 1 packets: 54
      14.9999455 in 2 packets: 72, 108
      7.9999708 in 1 packets: 90
Temperature FPGA: 
      04 -> 49.8 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 91, 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
      05 -> 48.5 degC in 18 packets: 109, 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 89 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 89 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20230518_102637_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.5475379 to 9.5478269
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.7457110  -- OK for SL --
   MIU time from 2023-05-18T10:26:47.785.955.000 to 2023-05-18T10:26:47.786.244.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.745772

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.2935379 to 26.2938269
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.1976310  -- OK for SL --
   MIU time from 2023-05-18T10:27:04.532.016.000 to 2023-05-18T10:27:04.532.305.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.197660

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.4914579 to 34.4917469
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.2343350  -- OK for SL --
   MIU time from 2023-05-18T10:27:12.729.965.000 to 2023-05-18T10:27:12.730.254.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.234412

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.7260819 to 55.7263709
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0057110  -- OK for SL --
   MIU time from 2023-05-18T10:27:33.964.666.000 to 2023-05-18T10:27:33.964.955.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.005766

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 70.7320819 to 70.7323709
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.2479030  -- OK for SL --
   MIU time from 2023-05-18T10:27:48.970.721.000 to 2023-05-18T10:27:48.971.010.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.247931

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 77.9802739 to 77.9805629
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8358390  -- OK for SL --
   MIU time from 2023-05-18T10:27:56.218.941.000 to 2023-05-18T10:27:56.219.230.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.835894

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 92.8164019 to 92.8166909
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-18T10:28:11.055.124.000 to 2023-05-18T10:28:11.055.413.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.547537900 corrected by 40bit 10MHz counter of 8.761582700 obtained at 1Hz counter of 7. Result: 7.785958108
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2023-05-18T10:26:47.785.814.725
   Max 1st sample MIU time among all SL packets: 2023-05-18T10:28:11.055.272.725
   Min 1st sample XO time among all SL packets: 7.785817833
   Max 1st sample XO time among all SL packets: 91.055273824
   TM values from -254 to 230 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-18T10:26:47.785.814.725 (XO: 7.785817833 - 7.786123834, t=37.2 degC)  Rank: 00089
   TM values: Max =   215  Min =  -230  Ave =    -4.27  Std =    20.60
   Min TM value:-230 at an offset of 117.463 usec -> 2023-05-18T10:26:47.785.932.187 (XO: 7.785935295)
   Max TM value:215 at an offset of 117.700 usec -> 2023-05-18T10:26:47.785.932.425 (XO: 7.785935533)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 117.550 usec -> 2023-05-18T10:26:47.785.932.275 (XO: 7.785935383)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 119.638 usec -> 2023-05-18T10:26:47.785.934.362 (XO: 7.785937470)

 - Block number: 2 starting at 2023-05-18T10:27:04.531.875.725 (XO: 24.531878740 - 24.532184741, t=37.2 degC)  Rank: 00078
   TM values: Max =   174  Min =  -230  Ave =    -4.32  Std =    16.59
   Min TM value:-230 at an offset of 118.138 usec -> 2023-05-18T10:27:04.531.993.862 (XO: 24.531996878)
   Max TM value:174 at an offset of 117.838 usec -> 2023-05-18T10:27:04.531.993.562 (XO: 24.531996578)
   Max dTM/dt [1/us]:1.53e+03 at an offset of 120.650 usec -> 2023-05-18T10:27:04.531.996.375 (XO: 24.531999390)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 117.975 usec -> 2023-05-18T10:27:04.531.993.700 (XO: 24.531996715)

 - Block number: 3 starting at 2023-05-18T10:27:12.729.824.725 (XO: 32.729828525 - 32.730134526, t=37.2 degC)  Rank: 00084
   TM values: Max =   230  Min =  -254  Ave =    -4.29  Std =    29.20
   Min TM value:-254 at an offset of 119.475 usec -> 2023-05-18T10:27:12.729.944.200 (XO: 32.729948001)
   Max TM value:230 at an offset of 119.213 usec -> 2023-05-18T10:27:12.729.943.937 (XO: 32.729947738)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 117.250 usec -> 2023-05-18T10:27:12.729.941.975 (XO: 32.729945776)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 119.338 usec -> 2023-05-18T10:27:12.729.944.062 (XO: 32.729947863)

 - Block number: 4 starting at 2023-05-18T10:27:33.964.525.725 (XO: 53.964529897 - 53.964835898, t=37.2 degC)  Rank: 00076
   TM values: Max =   181  Min =  -231  Ave =    -4.35  Std =    18.57
   Min TM value:-231 at an offset of 106.113 usec -> 2023-05-18T10:27:33.964.631.837 (XO: 53.964636010)
   Max TM value:181 at an offset of 93.475 usec -> 2023-05-18T10:27:33.964.619.200 (XO: 53.964623373)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 108.763 usec -> 2023-05-18T10:27:33.964.634.487 (XO: 53.964638660)
   Min dTM/dt [1/us]:-1.91e+03 at an offset of 93.600 usec -> 2023-05-18T10:27:33.964.619.325 (XO: 53.964623498)

 - Block number: 5 starting at 2023-05-18T10:27:48.970.580.725 (XO: 68.970584516 - 68.970890517, t=37.2 degC)  Rank: 00079
   TM values: Max =   175  Min =  -219  Ave =    -4.25  Std =    17.22
   Min TM value:-219 at an offset of 122.863 usec -> 2023-05-18T10:27:48.970.703.587 (XO: 68.970707379)
   Max TM value:175 at an offset of 125.688 usec -> 2023-05-18T10:27:48.970.706.412 (XO: 68.970710204)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 122.975 usec -> 2023-05-18T10:27:48.970.703.700 (XO: 68.970707492)
   Min dTM/dt [1/us]:-1.62e+03 at an offset of 120.225 usec -> 2023-05-18T10:27:48.970.700.950 (XO: 68.970704742)

 - Block number: 6 starting at 2023-05-18T10:27:56.218.800.725 (XO: 76.218802913 - 76.219108914, t=37.2 degC)  Rank: 00079
   TM values: Max =   193  Min =  -233  Ave =    -4.36  Std =    20.22
   Min TM value:-233 at an offset of 117.638 usec -> 2023-05-18T10:27:56.218.918.362 (XO: 76.218920551)
   Max TM value:193 at an offset of 117.375 usec -> 2023-05-18T10:27:56.218.918.100 (XO: 76.218920288)
   Max dTM/dt [1/us]:1.49e+03 at an offset of 120.100 usec -> 2023-05-18T10:27:56.218.920.825 (XO: 76.218923013)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 117.475 usec -> 2023-05-18T10:27:56.218.918.200 (XO: 76.218920388)

 - Block number: 7 starting at 2023-05-18T10:28:11.054.983.725 (XO: 91.054984823 - 91.055290824, t=37.2 degC)  Rank: 00078
   TM values: Max =   193  Min =  -233  Ave =    -4.37  Std =    19.74
   Min TM value:-233 at an offset of 119.988 usec -> 2023-05-18T10:28:11.055.103.712 (XO: 91.055104811)
   Max TM value:193 at an offset of 119.713 usec -> 2023-05-18T10:28:11.055.103.437 (XO: 91.055104536)
   Max dTM/dt [1/us]:1.60e+03 at an offset of 117.650 usec -> 2023-05-18T10:28:11.055.101.375 (XO: 91.055102473)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 119.813 usec -> 2023-05-18T10:28:11.055.103.537 (XO: 91.055104636)
