-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Thu May 18 22:08:17 2023 UTC
-- File 20230518_100112_1229.dat, 1901  packets found.
APID:
      1229 (EW) in 1901 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 1901 packets (all packets)
Total number of packets: 
      36720 in 1901 packets (all packets)
Packet No from 1 to 1901
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 77.546731 to 77.546731
-each separately from 77s to 77s and from 546731us to 546731us.
   Differences in neigboring packets: 
      0.000000 in 1900 packets (all packets)
MIU packet time from 2023-05-18T10:02:32.546.731.000 to 2023-05-18T10:02:32.546.731.000
   Differences in neigboring packets: 
      0.000000 in 1900 packets (all packets)
MIU orbit number: 
      60000 in 1901 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 1901 packets (all packets)
MIU orbit information / number of events: 
      0 in 1901 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 1901 packets (all packets)
Instrument anomalies: 
      0 (OK) in 1901 packets (all packets)
FPGA configuration version: 
      211 in 1901 packets (all packets)
10 MHz counter at the last 1Hz pulse from 78.8719118 to 78.8719118
   Differences in neigboring packets: 
      0.0000000 in 1900 packets (all packets)
Temperature FPGA: 
      05 -> 48.5 degC in 1901 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 1901 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 1901 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 1901 packets (all packets)
Number of external events: 
      2 in 1901 packets (all packets)
Number of generated alerts: 
      2 in 1901 packets (all packets)
Survey memory error: 
      0 in 1901 packets (all packets)
Event memory error: 
      0 in 1901 packets (all packets)
Configuration error: 
      0 in 1901 packets (all packets)
SDRAM CRC error: 
      0 in 1901 packets (all packets)
Radio gain: 
      0 in 1901 packets (all packets)
Radio bandwidth: 
      0 in 1901 packets (all packets)
Digital gain: 
      0 in 1901 packets (all packets)
Analog gain: 
      0 in 1901 packets (all packets)

EVENT DATA READER
-- File 20230518_100112_1229.dat, 9 EW pages found. Extra packets: 65.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 2 to 2
      2 in 1836 packets (all packets)
   LS 32 bits of C10MHz from descriptor (period 430s)  from 79.3631559 to 79.3943509
      Difference between the neighboring packets: 
      0.0000170 in 1835 packets (all packets)
   Memory page number from 33 to 41
      Difference between the neighboring packets: 
      0 in 1827 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 8 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632
   Event page number from 1 to 9
      Difference between the neighboring packets: 
      0 in 1827 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 8 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632
   MIU event flag 
      1 in 1836 packets (all packets)
   Time triggered event flag 
      0 in 1836 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2023-05-18T10:02:32.491.242.620
   Min EW header abs. time found among all events            : 2023-05-18T10:02:32.546.731.000
   Max EW event (1st sample) abs. time found among all events: 2023-05-18T10:02:32.491.242.620
   Max EW header abs. time found among all events            : 2023-05-18T10:02:32.546.731.000
   TM values from -224 to 195 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 2 total number of packets: 1836
    Page no. 1:  204 packets  -- OK;  memory page(s):
      33 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      34 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      35 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      36 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      37 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      38 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      39 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      40 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      41 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 9  ordered packets: 1836
   TM values from -224 to 195 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=77 C1MHz=0.546731
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 1835 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 79.3631559
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-05-18T10:02:32.546.731.000
   Event 1st sample abs. time : 2023-05-18T10:02:32.491.242.620)
   --- Rel. time correction for 1 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 77, 32bit 10MHz counter = 79.363155900 corrected by 40bit 10MHz counter of 74.871926400 obtained at 1Hz counter of 73. Result: 77.491246118
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              77.491245943
   Event data descriptor XO time (reconstructed seconds from XP start)    :              77.491246118
   Sampling frequency correction factor for this event:     0.999996300
--
Extremes in separate events:

 - Event number: 1 --EMPTY-- 

 - Event number: 2 from 2023-05-18T10:02:32.491.242.620 to 2023-05-18T10:02:32.522.454.735 (XO: 77.491245943 - 77.522458058)
   Min TM value:-224 at an offset of 22958.397 usec -> 2023-05-18T10:02:32.514.201.017 (XO: 77.514204340)
   Max TM value:195 at an offset of 22958.160 usec -> 2023-05-18T10:02:32.514.200.779 (XO: 77.514204103)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 22955.972 usec -> 2023-05-18T10:02:32.514.198.592 (XO: 77.514201915)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 22958.285 usec -> 2023-05-18T10:02:32.514.200.904 (XO: 77.514204228)
   Med: -4.00000   Ave: -4.4172698   Std: 4.7994254
