-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri May 12 22:47:39 2023 UTC
-- File 20230512_093235_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.850133 to 18.132687
-each separately from 7s to 18s and from 132398us to 850422us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.281976 in 1 packets: 18
MIU packet time from 2023-05-12T09:32:45.850.133.000 to 2023-05-12T09:32:56.132.687.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.281976 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.9235480 to 19.9235075
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999595 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 36 packets (all packets)
Number of generated alerts: 
      1 in 3 packets: 1, 2, 3
      2 in 6 packets: 4, 5, 6, 7, 8, 9
      3 in 27 packets: 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, 20, 21, 22, 23, 24, 25, 26, 27, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230512_093235_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      167 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.7736822 to 9.7739712
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.2819350  -- OK for SL --
   MIU time from 2023-05-12T09:32:45.850.133.000 to 2023-05-12T09:32:45.850.422.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.281976

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      52 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.0559062 to 20.0561952
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-12T09:32:56.132.398.000 to 2023-05-12T09:32:56.132.687.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.773682200 corrected by 40bit 10MHz counter of 8.923548000 obtained at 1Hz counter of 7. Result: 7.850137260
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2023-05-12T09:32:45.849.992.725
   Max 1st sample MIU time among all SL packets: 2023-05-12T09:32:56.132.546.725
   Min 1st sample XO time among all SL packets: 7.849996985
   Max 1st sample XO time among all SL packets: 18.132547915
   TM values from -725 to 872 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-12T09:32:45.849.992.725 (XO: 7.849996985 - 7.850302986, t=38.8 degC)  Rank: 00167
   TM values: Max =   872  Min =  -725  Ave =    -4.35  Std =   126.74
   Min TM value:-725 at an offset of 171.738 usec -> 2023-05-12T09:32:45.850.164.463 (XO: 7.850168723)
   Max TM value:872 at an offset of 171.951 usec -> 2023-05-12T09:32:45.850.164.675 (XO: 7.850168936)
   Max dTM/dt [1/us]:9.51e+03 at an offset of 171.801 usec -> 2023-05-12T09:32:45.850.164.525 (XO: 7.850168786)
   Min dTM/dt [1/us]:-7.46e+03 at an offset of 172.038 usec -> 2023-05-12T09:32:45.850.164.763 (XO: 7.850169023)

 - Block number: 2 starting at 2023-05-12T09:32:56.132.257.725 (XO: 18.132258914 - 18.132564915, t=38.8 degC)  Rank: 00052
   TM values: Max =   157  Min =  -147  Ave =    -4.45  Std =    12.79
   Min TM value:-147 at an offset of 112.663 usec -> 2023-05-12T09:32:56.132.370.387 (XO: 18.132371577)
   Max TM value:157 at an offset of 100.163 usec -> 2023-05-12T09:32:56.132.357.887 (XO: 18.132359077)
   Max dTM/dt [1/us]:1.26e+03 at an offset of 100.075 usec -> 2023-05-12T09:32:56.132.357.800 (XO: 18.132358990)
   Min dTM/dt [1/us]:-1.35e+03 at an offset of 100.275 usec -> 2023-05-12T09:32:56.132.358.000 (XO: 18.132359190)
