-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri May 12 22:26:37 2023 UTC
-- File 20230512_082807_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177031
-each separately from 0s to 14s and from 0us to 841256us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 9 packets: 2, 3, 5, 9, 10, 12, 14, 15, 17
      0.833943 in 5 packets: 4, 6, 8, 11, 16
      0.833941 in 1 packets: 7
      0.833940 in 1 packets: 13
MIU packet time from 2023-05-12T08:28:10.000.000.000 to 2023-05-12T08:28:24.177.031.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 9 packets: 2, 3, 5, 9, 10, 12, 14, 15, 17
      0.833943 in 5 packets: 4, 6, 8, 11, 16
      0.833941 in 1 packets: 7
      0.833940 in 1 packets: 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7254859 to 15.7254342
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999963 in 11 packets: 2, 6, 8, 9, 10, 11, 12, 14, 15, 16, 17
      0.9999964 in 2 packets: 3, 5
      0.9999962 in 1 packets: 4
   XO PPM: 
      -3.7 in 11 packets: 2, 6, 8, 9, 10, 11, 12, 14, 15, 16, 17
      -3.6 in 2 packets: 3, 5
      -3.8 in 1 packets: 4
Temperature FPGA: 
      03 -> 51.1 degC in 14 packets: 1, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 15, 16, 18
      04 -> 49.8 degC in 4 packets: 2, 13, 14, 17
Temperature XO: 
      C2 -> 38.8 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 13 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13
      2 in 5 packets: 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 13 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13
      3 in 5 packets: 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230512_082807_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.7254860 to 15.9024660
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-12T08:28:10.000.000.000 to 2023-05-12T08:28:24.177.031.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 9 packets: 2, 3, 5, 9, 10, 12, 14, 15, 17
      0.833943 in 5 packets: 4, 6, 8, 11, 16
      0.833941 in 1 packets: 7
      0.833940 in 1 packets: 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.725486000 corrected by 40bit 10MHz counter of 1.725485900 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996200 - 0.999996400
   Min 1st sample MIU time among all SR packets: 2023-05-12T08:28:09.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-05-12T08:28:24.177.030.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177032380
   TM values from -34 to 32 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-05-12T08:28:09.999.999.925 (XO: 0.000000025 - 0.000017025, t=38.8 degC)
   TM values: Max =    22  Min =   -31  Ave =    -4.19  Std =    15.07

 - Packet number: 2 starting at 2023-05-12T08:28:10.833.938.925 (XO: 0.833943111 - 0.833960111, t=38.8 degC)
   TM values: Max =    18  Min =   -28  Ave =    -4.65  Std =    11.76

 - Packet number: 3 starting at 2023-05-12T08:28:11.667.882.925 (XO: 1.667886129 - 1.667903129, t=38.8 degC)
   TM values: Max =    15  Min =   -27  Ave =    -4.33  Std =    10.09

 - Packet number: 4 starting at 2023-05-12T08:28:12.501.826.925 (XO: 2.501829232 - 2.501846232, t=38.8 degC)
   TM values: Max =    15  Min =   -22  Ave =    -4.20  Std =     9.85

 - Packet number: 5 starting at 2023-05-12T08:28:13.335.769.925 (XO: 3.335772334 - 3.335789334, t=38.8 degC)
   TM values: Max =    12  Min =   -22  Ave =    -4.41  Std =     8.10

 - Packet number: 6 starting at 2023-05-12T08:28:14.169.713.925 (XO: 4.169715353 - 4.169732353, t=38.8 degC)
   TM values: Max =    21  Min =   -30  Ave =    -4.41  Std =    14.68

 - Packet number: 7 starting at 2023-05-12T08:28:15.003.656.925 (XO: 5.003658439 - 5.003675439, t=38.8 degC)
   TM values: Max =    15  Min =   -23  Ave =    -4.21  Std =     9.85

 - Packet number: 8 starting at 2023-05-12T08:28:15.837.597.925 (XO: 5.837601524 - 5.837618524, t=38.8 degC)
   TM values: Max =    16  Min =   -27  Ave =    -4.54  Std =    10.67

 - Packet number: 9 starting at 2023-05-12T08:28:16.671.540.925 (XO: 6.671544610 - 6.671561610, t=38.8 degC)
   TM values: Max =    16  Min =   -25  Ave =    -4.29  Std =    10.05

 - Packet number: 10 starting at 2023-05-12T08:28:17.505.484.925 (XO: 7.505487695 - 7.505504695, t=38.8 degC)
   TM values: Max =    26  Min =   -33  Ave =    -4.40  Std =    16.85

 - Packet number: 11 starting at 2023-05-12T08:28:18.339.428.925 (XO: 8.339430781 - 8.339447781, t=38.8 degC)
   TM values: Max =    32  Min =   -34  Ave =    -4.46  Std =    15.51

 - Packet number: 12 starting at 2023-05-12T08:28:19.173.371.925 (XO: 9.173373866 - 9.173390867, t=38.8 degC)
   TM values: Max =    15  Min =   -25  Ave =    -4.07  Std =    10.45

 - Packet number: 13 starting at 2023-05-12T08:28:20.007.315.925 (XO: 10.007316952 - 10.007333952, t=38.8 degC)
   TM values: Max =    17  Min =   -27  Ave =    -4.62  Std =    11.82

 - Packet number: 14 starting at 2023-05-12T08:28:20.841.255.925 (XO: 10.841260038 - 10.841277038, t=38.8 degC)
   TM values: Max =    11  Min =   -21  Ave =    -4.39  Std =     7.37

 - Packet number: 15 starting at 2023-05-12T08:28:21.675.199.925 (XO: 11.675203123 - 11.675220123, t=38.8 degC)
   TM values: Max =    21  Min =   -30  Ave =    -4.55  Std =    13.76

 - Packet number: 16 starting at 2023-05-12T08:28:22.509.143.925 (XO: 12.509146209 - 12.509163209, t=38.8 degC)
   TM values: Max =    21  Min =   -28  Ave =    -4.37  Std =    13.70

 - Packet number: 17 starting at 2023-05-12T08:28:23.343.086.925 (XO: 13.343089294 - 13.343106294, t=38.8 degC)
   TM values: Max =    13  Min =   -22  Ave =    -4.35  Std =     8.44

 - Packet number: 18 starting at 2023-05-12T08:28:24.177.030.925 (XO: 14.177032380 - 14.177049380, t=38.8 degC)
   TM values: Max =    14  Min =   -20  Ave =    -4.21  Std =     6.73
