-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri May 12 21:41:02 2023 UTC
-- File 20230512_043029_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177030
-each separately from 0s to 14s and from 0us to 841256us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 7 packets: 2, 4, 5, 9, 11, 14, 16
      0.833943 in 7 packets: 3, 6, 8, 10, 12, 15, 17
      0.833941 in 2 packets: 7, 13
MIU packet time from 2023-05-12T04:30:32.000.000.000 to 2023-05-12T04:30:46.177.030.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 7 packets: 2, 4, 5, 9, 11, 14, 16
      0.833943 in 7 packets: 3, 6, 8, 10, 12, 15, 17
      0.833941 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8093117 to 15.8092606
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999964 in 7 packets: 2, 4, 6, 10, 12, 15, 17
      0.9999963 in 7 packets: 3, 5, 8, 9, 11, 14, 16
   XO PPM: 
      -3.6 in 7 packets: 2, 4, 6, 10, 12, 15, 17
      -3.7 in 7 packets: 3, 5, 8, 9, 11, 14, 16
Temperature FPGA: 
      03 -> 51.1 degC in 8 packets: 1, 3, 4, 6, 10, 11, 12, 16
      04 -> 49.8 degC in 10 packets: 2, 5, 7, 8, 9, 13, 14, 15, 17, 18
Temperature XO: 
      C2 -> 38.8 degC in 4 packets: 1, 7, 8, 15
      C1 -> 37.2 degC in 14 packets: 2, 3, 4, 5, 6, 9, 10, 11, 12, 13, 14, 16, 17, 18
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 12 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12
      1 in 6 packets: 13, 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 12 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12
      1 in 6 packets: 13, 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230512_043029_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.8093118 to 15.9862918
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-12T04:30:32.000.000.000 to 2023-05-12T04:30:46.177.030.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 7 packets: 2, 4, 5, 9, 11, 14, 16
      0.833943 in 7 packets: 3, 6, 8, 10, 12, 15, 17
      0.833941 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.809311800 corrected by 40bit 10MHz counter of 1.809311700 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SR packets: 2023-05-12T04:30:31.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-05-12T04:30:46.177.029.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177031762
   TM values from -48 to 37 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-05-12T04:30:31.999.999.925 (XO: 0.000000025 - 0.000017025, t=38.8 degC)
   TM values: Max =    29  Min =   -36  Ave =    -4.48  Std =    13.41

 - Packet number: 2 starting at 2023-05-12T04:30:32.833.938.925 (XO: 0.833943027 - 0.833960027, t=37.2 degC)
   TM values: Max =    12  Min =   -21  Ave =    -4.23  Std =     5.98

 - Packet number: 3 starting at 2023-05-12T04:30:33.667.882.925 (XO: 1.667886096 - 1.667903096, t=37.2 degC)
   TM values: Max =    16  Min =   -26  Ave =    -4.44  Std =     7.97

 - Packet number: 4 starting at 2023-05-12T04:30:34.501.825.925 (XO: 2.501829132 - 2.501846132, t=37.2 degC)
   TM values: Max =    20  Min =   -30  Ave =    -4.31  Std =    11.12

 - Packet number: 5 starting at 2023-05-12T04:30:35.335.769.925 (XO: 3.335772167 - 3.335789167, t=37.2 degC)
   TM values: Max =    11  Min =   -20  Ave =    -4.53  Std =     5.10

 - Packet number: 6 starting at 2023-05-12T04:30:36.169.713.925 (XO: 4.169715236 - 4.169732236, t=37.2 degC)
   TM values: Max =    35  Min =   -44  Ave =    -4.44  Std =    21.07

 - Packet number: 7 starting at 2023-05-12T04:30:37.003.656.925 (XO: 5.003658239 - 5.003675239, t=38.8 degC)
   TM values: Max =    35  Min =   -44  Ave =    -4.47  Std =    21.92

 - Packet number: 8 starting at 2023-05-12T04:30:37.837.597.925 (XO: 5.837601324 - 5.837618324, t=38.8 degC)
   TM values: Max =    33  Min =   -40  Ave =    -4.50  Std =    17.76

 - Packet number: 9 starting at 2023-05-12T04:30:38.671.540.925 (XO: 6.671544410 - 6.671561410, t=37.2 degC)
   TM values: Max =    14  Min =   -23  Ave =    -4.56  Std =     6.08

 - Packet number: 10 starting at 2023-05-12T04:30:39.505.484.925 (XO: 7.505487445 - 7.505504445, t=37.2 degC)
   TM values: Max =    20  Min =   -29  Ave =    -4.40  Std =     9.06

 - Packet number: 11 starting at 2023-05-12T04:30:40.339.427.925 (XO: 8.339430481 - 8.339447481, t=37.2 degC)
   TM values: Max =    11  Min =   -19  Ave =    -4.31  Std =     5.27

 - Packet number: 12 starting at 2023-05-12T04:30:41.173.371.925 (XO: 9.173373549 - 9.173390549, t=37.2 degC)
   TM values: Max =    37  Min =   -40  Ave =    -4.65  Std =    17.58

 - Packet number: 13 starting at 2023-05-12T04:30:42.007.314.925 (XO: 10.007316552 - 10.007333552, t=37.2 degC)
   TM values: Max =    34  Min =   -48  Ave =    -4.50  Std =    20.28

 - Packet number: 14 starting at 2023-05-12T04:30:42.841.255.925 (XO: 10.841259638 - 10.841276638, t=37.2 degC)
   TM values: Max =    26  Min =   -35  Ave =    -4.34  Std =    13.62

 - Packet number: 15 starting at 2023-05-12T04:30:43.675.199.925 (XO: 11.675202656 - 11.675219656, t=38.8 degC)
   TM values: Max =    26  Min =   -35  Ave =    -4.29  Std =    14.47

 - Packet number: 16 starting at 2023-05-12T04:30:44.509.142.925 (XO: 12.509145709 - 12.509162709, t=37.2 degC)
   TM values: Max =    24  Min =   -31  Ave =    -4.30  Std =    11.78

 - Packet number: 17 starting at 2023-05-12T04:30:45.343.086.925 (XO: 13.343088760 - 13.343105760, t=37.2 degC)
   TM values: Max =    22  Min =   -31  Ave =    -4.38  Std =    11.44

 - Packet number: 18 starting at 2023-05-12T04:30:46.177.029.925 (XO: 14.177031762 - 14.177048762, t=37.2 degC)
   TM values: Max =    26  Min =   -31  Ave =    -4.48  Std =    12.19
