-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu May 11 23:10:49 2023 UTC
-- File 20230511_143252_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.056950 to 32.441436
-each separately from 13s to 32s and from 56950us to 541025us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.483497 in 1 packets: 18
      2.900122 in 1 packets: 36
MIU packet time from 2023-05-11T14:33:08.056.950.000 to 2023-05-11T14:33:27.441.436.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.483497 in 1 packets: 18
      2.900122 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.7255519 to 33.7254831
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999421 in 1 packets: 18
      2.9999891 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      C2 -> 38.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      1 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Number of generated alerts: 
      1 in 3 packets: 1, 2, 3
      2 in 15 packets: 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      7 in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20230511_143252_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      415 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.7825026 to 14.7827916
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.4834390  -- OK for SL --
   MIU time from 2023-05-11T14:33:08.056.950.000 to 2023-05-11T14:33:08.057.239.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.483497

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      66 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.2662306 to 31.2665196
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.9001110  -- OK for SL --
   MIU time from 2023-05-11T14:33:24.540.736.000 to 2023-05-11T14:33:24.541.025.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.900122

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      63 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.1666306 to 34.1669196
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-11T14:33:27.441.147.000 to 2023-05-11T14:33:27.441.436.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 14.782502600 corrected by 40bit 10MHz counter of 14.725551900 obtained at 1Hz counter of 13. Result: 13.056950905
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2023-05-11T14:33:08.056.809.725
   Max 1st sample MIU time among all SL packets: 2023-05-11T14:33:27.441.295.725
   Min 1st sample XO time among all SL packets: 13.056810630
   Max 1st sample XO time among all SL packets: 32.441297814
   TM values from -514 to 423 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-11T14:33:08.056.809.725 (XO: 13.056810630 - 13.057116631, t=37.2 degC)  Rank: 00415
   TM values: Max =   423  Min =  -514  Ave =    -4.57  Std =    98.40
   Min TM value:-514 at an offset of 90.375 usec -> 2023-05-11T14:33:08.056.900.100 (XO: 13.056901005)
   Max TM value:423 at an offset of 86.950 usec -> 2023-05-11T14:33:08.056.896.675 (XO: 13.056897580)
   Max dTM/dt [1/us]:2.41e+03 at an offset of 129.675 usec -> 2023-05-11T14:33:08.056.939.400 (XO: 13.056940305)
   Min dTM/dt [1/us]:-1.29e+03 at an offset of 142.951 usec -> 2023-05-11T14:33:08.056.952.675 (XO: 13.056953580)

 - Block number: 2 starting at 2023-05-11T14:33:24.540.595.725 (XO: 29.540598271 - 29.540904272, t=38.8 degC)  Rank: 00066
   TM values: Max =   183  Min =  -202  Ave =    -4.43  Std =    13.85
   Min TM value:-202 at an offset of 129.825 usec -> 2023-05-11T14:33:24.540.725.550 (XO: 29.540728097)
   Max TM value:183 at an offset of 131.888 usec -> 2023-05-11T14:33:24.540.727.612 (XO: 29.540730159)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 129.938 usec -> 2023-05-11T14:33:24.540.725.662 (XO: 29.540728209)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 132.000 usec -> 2023-05-11T14:33:24.540.727.725 (XO: 29.540730272)

 - Block number: 3 starting at 2023-05-11T14:33:27.441.006.725 (XO: 32.441008813 - 32.441314814, t=38.8 degC)  Rank: 00063
   TM values: Max =   197  Min =  -186  Ave =    -4.44  Std =    17.70
   Min TM value:-186 at an offset of 116.650 usec -> 2023-05-11T14:33:27.441.123.375 (XO: 32.441125463)
   Max TM value:197 at an offset of 118.788 usec -> 2023-05-11T14:33:27.441.125.512 (XO: 32.441127601)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 118.713 usec -> 2023-05-11T14:33:27.441.125.437 (XO: 32.441127526)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 118.875 usec -> 2023-05-11T14:33:27.441.125.600 (XO: 32.441127688)
