-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Apr 26 23:20:10 2023 UTC
-- File 20230426_162024_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.033559 to 77.454231
-each separately from 11s to 77s and from 33559us to 854189us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.820052 in 1 packets: 18
      17.429968 in 1 packets: 36
      12.429498 in 1 packets: 54
      9.919811 in 1 packets: 72
      15.819609 in 1 packets: 90
MIU packet time from 2023-04-26T16:20:38.033.559.000 to 2023-04-26T16:21:44.454.231.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.820052 in 1 packets: 18
      17.429968 in 1 packets: 36
      12.429498 in 1 packets: 54
      9.919811 in 1 packets: 72
      15.819609 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.2533268 to 79.2530885
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999639 in 1 packets: 18
      17.9999350 in 1 packets: 36
      11.9999567 in 1 packets: 54
      9.9999640 in 1 packets: 72
      15.9999421 in 1 packets: 90
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      03 -> 51.1 degC in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 71 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 71 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20230426_162024_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.2868873 to 13.2871763
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.8200150  -- OK for SL --
   MIU time from 2023-04-26T16:20:38.033.559.000 to 2023-04-26T16:20:38.033.848.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.820052

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.1071913 to 24.1074803
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.4299030  -- OK for SL --
   MIU time from 2023-04-26T16:20:48.853.900.000 to 2023-04-26T16:20:48.854.189.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.429968

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.5373833 to 41.5376723
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.4294550  -- OK for SL --
   MIU time from 2023-04-26T16:21:06.284.157.000 to 2023-04-26T16:21:06.284.446.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.429498

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 53.9671273 to 53.9674163
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.9197750  -- OK for SL --
   MIU time from 2023-04-26T16:21:18.713.944.000 to 2023-04-26T16:21:18.714.233.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.919811

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 63.8871913 to 63.8874803
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.8195510  -- OK for SL --
   MIU time from 2023-04-26T16:21:28.634.044.000 to 2023-04-26T16:21:28.634.333.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.819609

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 79.7070313 to 79.7073203
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-04-26T16:21:44.453.942.000 to 2023-04-26T16:21:44.454.231.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.286887300 corrected by 40bit 10MHz counter of 13.253326800 obtained at 1Hz counter of 11. Result: 11.033560621
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-04-26T16:20:38.033.418.725
   Max 1st sample MIU time among all SL packets: 2023-04-26T16:21:44.454.090.725
   Min 1st sample XO time among all SL packets: 11.033420345
   Max 1st sample XO time among all SL packets: 77.454093160
   TM values from -247 to 222 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-04-26T16:20:38.033.418.725 (XO: 11.033420345 - 11.033726346, t=37.2 degC)  Rank: 00077
   TM values: Max =   185  Min =  -239  Ave =    -4.42  Std =    21.44
   Min TM value:-239 at an offset of 85.238 usec -> 2023-04-26T16:20:38.033.503.962 (XO: 11.033505583)
   Max TM value:185 at an offset of 84.988 usec -> 2023-04-26T16:20:38.033.503.712 (XO: 11.033505333)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 82.800 usec -> 2023-04-26T16:20:38.033.501.525 (XO: 11.033503146)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 112.275 usec -> 2023-04-26T16:20:38.033.531.000 (XO: 11.033532621)

 - Block number: 2 starting at 2023-04-26T16:20:48.853.759.725 (XO: 21.853763399 - 21.854069400, t=37.2 degC)  Rank: 00082
   TM values: Max =   190  Min =  -226  Ave =    -4.47  Std =    18.62
   Min TM value:-226 at an offset of 123.875 usec -> 2023-04-26T16:20:48.853.883.600 (XO: 21.853887274)
   Max TM value:190 at an offset of 111.125 usec -> 2023-04-26T16:20:48.853.870.850 (XO: 21.853874524)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 126.550 usec -> 2023-04-26T16:20:48.853.886.275 (XO: 21.853889949)
   Min dTM/dt [1/us]:-1.87e+03 at an offset of 111.263 usec -> 2023-04-26T16:20:48.853.870.987 (XO: 21.853874661)

 - Block number: 3 starting at 2023-04-26T16:21:06.284.016.725 (XO: 39.284018347 - 39.284324349, t=37.2 degC)  Rank: 00081
   TM values: Max =   183  Min =  -246  Ave =    -4.52  Std =    18.85
   Min TM value:-246 at an offset of 115.013 usec -> 2023-04-26T16:21:06.284.131.737 (XO: 39.284133360)
   Max TM value:183 at an offset of 115.275 usec -> 2023-04-26T16:21:06.284.132.000 (XO: 39.284133623)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 115.125 usec -> 2023-04-26T16:21:06.284.131.850 (XO: 39.284133473)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 112.438 usec -> 2023-04-26T16:21:06.284.129.162 (XO: 39.284130785)

 - Block number: 4 starting at 2023-04-26T16:21:18.713.803.725 (XO: 51.713807123 - 51.714113124, t=37.2 degC)  Rank: 00077
   TM values: Max =   183  Min =  -247  Ave =    -4.50  Std =    21.76
   Min TM value:-247 at an offset of 97.225 usec -> 2023-04-26T16:21:18.713.900.950 (XO: 51.713904349)
   Max TM value:183 at an offset of 96.975 usec -> 2023-04-26T16:21:18.713.900.700 (XO: 51.713904099)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 126.463 usec -> 2023-04-26T16:21:18.713.930.187 (XO: 51.713933586)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 97.138 usec -> 2023-04-26T16:21:18.713.900.862 (XO: 51.713904261)

 - Block number: 5 starting at 2023-04-26T16:21:28.633.903.725 (XO: 61.633906907 - 61.634212908, t=37.2 degC)  Rank: 00075
   TM values: Max =   206  Min =  -243  Ave =    -4.41  Std =    19.14
   Min TM value:-243 at an offset of 87.300 usec -> 2023-04-26T16:21:28.633.991.025 (XO: 61.633994207)
   Max TM value:206 at an offset of 74.388 usec -> 2023-04-26T16:21:28.633.978.112 (XO: 61.633981295)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 89.838 usec -> 2023-04-26T16:21:28.633.993.562 (XO: 61.633996745)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 87.150 usec -> 2023-04-26T16:21:28.633.990.875 (XO: 61.633994057)

 - Block number: 6 starting at 2023-04-26T16:21:44.453.801.725 (XO: 77.453804159 - 77.454110160, t=37.2 degC)  Rank: 00073
   TM values: Max =   222  Min =  -224  Ave =    -4.43  Std =    19.30
   Min TM value:-224 at an offset of 87.900 usec -> 2023-04-26T16:21:44.453.889.625 (XO: 77.453892059)
   Max TM value:222 at an offset of 74.963 usec -> 2023-04-26T16:21:44.453.876.687 (XO: 77.453879121)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 90.438 usec -> 2023-04-26T16:21:44.453.892.162 (XO: 77.453894597)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 75.088 usec -> 2023-04-26T16:21:44.453.876.812 (XO: 77.453879246)
