-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Apr 23 22:49:06 2023 UTC
-- File 20230423_141547_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.570077 to 41.242038
-each separately from 1s to 41s and from 241749us to 801649us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.230994 in 1 packets: 18
      13.440100 in 1 packets: 36
MIU packet time from 2023-04-23T14:15:51.570.077.000 to 2023-04-23T14:16:31.242.038.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.230994 in 1 packets: 18
      13.440100 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.7611069 to 42.7609545
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.9999010 in 1 packets: 18
      13.9999466 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20230423_141547_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3311847 to 3.3314737
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.2308950  -- OK for SL --
   MIU time from 2023-04-23T14:15:51.570.077.000 to 2023-04-23T14:15:51.570.366.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.230994

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.5623687 to 29.5626577
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4400470  -- OK for SL --
   MIU time from 2023-04-23T14:16:17.801.360.000 to 2023-04-23T14:16:17.801.649.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.440100

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.0027047 to 43.0029937
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-04-23T14:16:31.241.749.000 to 2023-04-23T14:16:31.242.038.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.331184700 corrected by 40bit 10MHz counter of 2.761106900 obtained at 1Hz counter of 1. Result: 1.570079966
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996200 - 0.999996200
   Min 1st sample MIU time among all SL packets: 2023-04-23T14:15:51.569.936.725
   Max 1st sample MIU time among all SL packets: 2023-04-23T14:16:31.241.897.725
   Min 1st sample XO time among all SL packets: 1.569939691
   Max 1st sample XO time among all SL packets: 41.241899844
   TM values from -228 to 195 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-04-23T14:15:51.569.936.725 (XO: 1.569939691 - 1.570245692, t=37.2 degC)  Rank: 00075
   TM values: Max =   179  Min =  -223  Ave =    -4.52  Std =    23.41
   Min TM value:-223 at an offset of 125.738 usec -> 2023-04-23T14:15:51.570.062.462 (XO: 1.570065429)
   Max TM value:179 at an offset of 125.975 usec -> 2023-04-23T14:15:51.570.062.700 (XO: 1.570065666)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 125.850 usec -> 2023-04-23T14:15:51.570.062.575 (XO: 1.570065541)
   Min dTM/dt [1/us]:-1.66e+03 at an offset of 154.051 usec -> 2023-04-23T14:15:51.570.090.775 (XO: 1.570093741)

 - Block number: 2 starting at 2023-04-23T14:16:17.801.219.725 (XO: 27.801223570 - 27.801529571, t=37.2 degC)  Rank: 00073
   TM values: Max =   190  Min =  -216  Ave =    -4.44  Std =    15.55
   Min TM value:-216 at an offset of 125.025 usec -> 2023-04-23T14:16:17.801.344.750 (XO: 27.801348595)
   Max TM value:190 at an offset of 124.763 usec -> 2023-04-23T14:16:17.801.344.487 (XO: 27.801348333)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 127.575 usec -> 2023-04-23T14:16:17.801.347.300 (XO: 27.801351145)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 124.900 usec -> 2023-04-23T14:16:17.801.344.625 (XO: 27.801348470)

 - Block number: 3 starting at 2023-04-23T14:16:31.241.608.725 (XO: 41.241610843 - 41.241916844, t=37.2 degC)  Rank: 00083
   TM values: Max =   195  Min =  -228  Ave =    -4.49  Std =    18.23
   Min TM value:-228 at an offset of 124.225 usec -> 2023-04-23T14:16:31.241.732.950 (XO: 41.241735069)
   Max TM value:195 at an offset of 122.238 usec -> 2023-04-23T14:16:31.241.730.962 (XO: 41.241733081)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 126.888 usec -> 2023-04-23T14:16:31.241.735.612 (XO: 41.241737731)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 124.113 usec -> 2023-04-23T14:16:31.241.732.837 (XO: 41.241734956)
