-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Sat Apr 22 21:52:00 2023 UTC
-- File 20230422_180404_1224.dat, 62  packets found.
APID:
      1224 (INIT) in 62 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 62 packets (all packets)
Total number of packets: 
      1 in 62 packets (all packets)
Packet No from 1 to 1
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 0.002092
-each separately from 0s to 0s and from 0us to 2092us.
   Differences in neigboring packets: 
      0.002092 in 1 packets: 1
      0.000000 in 60 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
MIU packet time from 2023-04-22T18:04:07.000.000.000 to 2023-04-22T18:04:07.002.092.000
   Differences in neigboring packets: 
      0.002092 in 1 packets: 1
      0.000000 in 60 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
MIU orbit number: 
      60000 in 62 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 62 packets (all packets)
MIU orbit information / number of events: 
      0 in 62 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 62 packets (all packets)
Instrument anomalies: 
      0 (OK) in 62 packets (all packets)
FPGA configuration version: 
      211 in 62 packets (all packets)
10 MHz counter at the last 1Hz pulse from 0.0000000 to 0.0000000
   Differences in neigboring packets: 
      0.0000000 in 61 packets (all packets)
Temperature FPGA: 
      00 -> 55.0 degC in 62 packets (all packets)
Temperature XO: 
      00 -> -273.1 degC in 62 packets (all packets)
Temperature ant 3: 
      00 -> 115.9 degC in 62 packets (all packets)
Temperature ant 4: 
      00 -> 115.9 degC in 62 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 61 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 62 packets (all packets)
Survey memory error: 
      0 in 62 packets (all packets)
Event memory error: 
      0 in 62 packets (all packets)
Configuration error: 
      0 in 1 packets: 1
      1 in 61 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      -> ERROR
SDRAM CRC error: 
      0 in 62 packets (all packets)
Radio gain: 
      0 in 62 packets (all packets)
Radio bandwidth: 
      0 in 62 packets (all packets)
Digital gain: 
      0 in 62 packets (all packets)
Analog gain: 
      12 in 62 packets (all packets)

Configuration from the INI packet written to 20230422_180404_1224.asc

Configuration check: mode validity (other than 0, 14 or 15): OK
Configuration check: completeness of CONFHFA (19 bytes)    : OK
Configuration check: completeness of HFPE/RW (19+120 bytes): OK
Configuration check: number of received CONFHFA bytes      : 19
Configuration check: number of HFPE/RW intervals (max. 20) : 20

Stack1 memory test: writing 0: OK    writing 1: OK
Stack2 memory test: writing 0: OK    writing 1: OK

SDRAM memory test: OK
Addr  0187A220 03447330 02E64AA0 01956FF0 015FD800 01F03400 00841700 00630E40 022944B0 039EF370 0128C560
      0128C560 03BCA7D0 0062F430 00538E20 027A4930 03476DA0 00E4DB70 0096B6C0 00DDEDA0 02B31B70 01F54B60
      01F54B60 0387F760 00440CD0 02660AB0 03550FE0 00FF8810 02804C10 01C06A10 03205F10 00B07090 02E848D0
      02E848D0 00CE3650 015496B0 01FEDDE0 0301B310 00816A90 02C1DFD0 01A13030 0171A820 03C97C30 022DC220
      022DC220 013B2330 02D35950 00DD7AF0 00B3C780 00EA2440 009F3660 02D0AD50 01B8FBF0 01648600 01D6C500
      01D6C500 013DA780 01A37440 02B96730 03F2EA50 000B9F70 000E50C0 000978A0 020DC4F0 030B2680 028EB5C0
      028EB5C0 03C9EF20 002D18B0 003B94E0 03132F40 034D5C70 02EBF240 039E0B60 00510ED0 027989B0 03454D60
      03454D60 00E7EBD0 02941E30 03DE1120 003119B0 0214CAB0 038F57F0 0248FC00 036C8200 02DAC300 03B7A280
      03B7A280 026C73C0 035A4A20 00F76F30 008CD8A0 02CAB4F0 01D7F740 029E0670 03D10540 023987E0 01254410
      01254410 03B7E610 006C1510
     

Error reading configuration from the INI packet:
  number of packets=62, APID=1224
