-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Apr  8 17:57:49 2023 UTC
-- File 20230407_145556_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.172352 to 9.172641
-each separately from 9s to 9s and from 172352us to 172641us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2023-04-07T14:56:08.172.352.000 to 2023-04-07T14:56:08.172.641.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8515285 to 10.8515285
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets (all packets)
Temperature XO: 
      BF -> 34.0 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230407_145556_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.0238814 to 11.0241704
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-04-07T14:56:08.172.352.000 to 2023-04-07T14:56:08.172.641.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.023881400 corrected by 40bit 10MHz counter of 10.851528500 obtained at 1Hz counter of 9. Result: 9.172353503
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-04-07T14:56:08.172.211.725
   Max 1st sample MIU time among all SL packets: 2023-04-07T14:56:08.172.500.725
   Min 1st sample XO time among all SL packets: 9.172213228
   Max 1st sample XO time among all SL packets: 9.172502229
   TM values from -228 to 191 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-04-07T14:56:08.172.211.725 (XO: 9.172213228 - 9.172519229, t=34.0 degC)  Rank: 00075
   TM values: Max =   191  Min =  -228  Ave =    -4.43  Std =    17.19
   Min TM value:-228 at an offset of 112.588 usec -> 2023-04-07T14:56:08.172.324.312 (XO: 9.172325816)
   Max TM value:191 at an offset of 112.350 usec -> 2023-04-07T14:56:08.172.324.075 (XO: 9.172325578)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 40.338 usec -> 2023-04-07T14:56:08.172.252.062 (XO: 9.172253565)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 112.463 usec -> 2023-04-07T14:56:08.172.324.187 (XO: 9.172325691)
