-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Apr  8 17:33:40 2023 UTC
-- File 20230407_141303_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.599712 to 17.548805
-each separately from 7s to 17s and from 548516us to 600001us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.948515 in 1 packets: 18
MIU packet time from 2023-04-07T14:13:13.599.712.000 to 2023-04-07T14:13:23.548.805.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.948515 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.4312530 to 18.4312175
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999645 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      BF -> 34.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230407_141303_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.0309656 to 9.0312546
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.9484790  -- OK for SL --
   MIU time from 2023-04-07T14:13:13.599.712.000 to 2023-04-07T14:13:13.600.001.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.948515

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      69 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.9797336 to 18.9800226
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-04-07T14:13:23.548.516.000 to 2023-04-07T14:13:23.548.805.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.030965600 corrected by 40bit 10MHz counter of 8.431253000 obtained at 1Hz counter of 7. Result: 7.599714759
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2023-04-07T14:13:13.599.571.725
   Max 1st sample MIU time among all SL packets: 2023-04-07T14:13:23.548.664.725
   Min 1st sample XO time among all SL packets: 7.599574483
   Max 1st sample XO time among all SL packets: 17.548666800
   TM values from -222 to 184 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-04-07T14:13:13.599.571.725 (XO: 7.599574483 - 7.599880485, t=34.0 degC)  Rank: 00072
   TM values: Max =   171  Min =  -222  Ave =    -4.51  Std =    16.95
   Min TM value:-222 at an offset of 114.038 usec -> 2023-04-07T14:13:13.599.685.762 (XO: 7.599688521)
   Max TM value:171 at an offset of 116.825 usec -> 2023-04-07T14:13:13.599.688.550 (XO: 7.599691309)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 116.700 usec -> 2023-04-07T14:13:13.599.688.425 (XO: 7.599691184)
   Min dTM/dt [1/us]:-1.68e+03 at an offset of 113.913 usec -> 2023-04-07T14:13:13.599.685.637 (XO: 7.599688396)

 - Block number: 2 starting at 2023-04-07T14:13:23.548.375.725 (XO: 17.548377799 - 17.548683800, t=34.0 degC)  Rank: 00069
   TM values: Max =   184  Min =  -222  Ave =    -4.51  Std =    15.84
   Min TM value:-222 at an offset of 116.213 usec -> 2023-04-07T14:13:23.548.491.937 (XO: 17.548494012)
   Max TM value:184 at an offset of 115.938 usec -> 2023-04-07T14:13:23.548.491.662 (XO: 17.548493737)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 118.763 usec -> 2023-04-07T14:13:23.548.494.487 (XO: 17.548496562)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 116.050 usec -> 2023-04-07T14:13:23.548.491.775 (XO: 17.548493850)
