-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Apr  4 00:34:38 2023 UTC
-- File 20230404_002529_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.533795 to 83.997536
-each separately from 5s to 83s and from 533795us to 997536us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.122996 in 1 packets: 18
      13.209217 in 1 packets: 36
      12.017181 in 1 packets: 54
      19.815291 in 1 packets: 72
      11.297322 in 1 packets: 90
MIU packet time from 2023-04-04T00:25:37.533.795.000 to 2023-04-04T00:26:55.997.536.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.122996 in 1 packets: 18
      13.209217 in 1 packets: 36
      12.017181 in 1 packets: 54
      19.815291 in 1 packets: 72
      11.297322 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.2533671 to 85.2530679
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999159 in 1 packets: 18
      12.9999501 in 1 packets: 36
      11.9999541 in 1 packets: 54
      19.9999231 in 1 packets: 72
      10.9999576 in 1 packets: 90
Temperature FPGA: 
      06 -> 47.2 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 36 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Temperature XO: 
      BF -> 34.0 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 37 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      3 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20230404_002529_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      67 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.7871629 to 7.7874519
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.1229110  -- OK for SL --
   MIU time from 2023-04-04T00:25:37.533.795.000 to 2023-04-04T00:25:37.534.084.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.122996

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      114 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.9103629 to 29.9106519
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.2091670  -- OK for SL --
   MIU time from 2023-04-04T00:25:59.657.080.000 to 2023-04-04T00:25:59.657.369.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.209217

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.1198189 to 43.1201079
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.0171350  -- OK for SL --
   MIU time from 2023-04-04T00:26:12.866.586.000 to 2023-04-04T00:26:12.866.875.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.017181

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.1372429 to 55.1375319
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.8152150  -- OK for SL --
   MIU time from 2023-04-04T00:26:24.884.056.000 to 2023-04-04T00:26:24.884.345.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.815291

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 74.9527469 to 74.9530359
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.2972790  -- OK for SL --
   MIU time from 2023-04-04T00:26:44.699.636.000 to 2023-04-04T00:26:44.699.925.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.297322

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 86.2503149 to 86.2506039
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-04-04T00:26:55.997.247.000 to 2023-04-04T00:26:55.997.536.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.787162900 corrected by 40bit 10MHz counter of 7.253367100 obtained at 1Hz counter of 5. Result: 5.533797828
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996100 - 0.999996200
   Min 1st sample MIU time among all SL packets: 2023-04-04T00:25:37.533.654.725
   Max 1st sample MIU time among all SL packets: 2023-04-04T00:26:55.997.395.725
   Min 1st sample XO time among all SL packets: 5.533657553
   Max 1st sample XO time among all SL packets: 83.997399615
   TM values from -828 to 409 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-04-04T00:25:37.533.654.725 (XO: 5.533657553 - 5.533963554, t=34.0 degC)  Rank: 00067
   TM values: Max =   190  Min =  -209  Ave =    -4.57  Std =    19.00
   Min TM value:-209 at an offset of 116.700 usec -> 2023-04-04T00:25:37.533.771.425 (XO: 5.533774253)
   Max TM value:190 at an offset of 116.425 usec -> 2023-04-04T00:25:37.533.771.150 (XO: 5.533773978)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 119.275 usec -> 2023-04-04T00:25:37.533.774.000 (XO: 5.533776828)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 116.538 usec -> 2023-04-04T00:25:37.533.771.262 (XO: 5.533774091)

 - Block number: 2 starting at 2023-04-04T00:25:59.656.939.725 (XO: 27.656942121 - 27.657248123, t=34.0 degC)  Rank: 00114
   TM values: Max =   409  Min =  -828  Ave =    -4.40  Std =    37.92
   Min TM value:-828 at an offset of 126.288 usec -> 2023-04-04T00:25:59.657.066.012 (XO: 27.657068409)
   Max TM value:409 at an offset of 127.025 usec -> 2023-04-04T00:25:59.657.066.750 (XO: 27.657069147)
   Max dTM/dt [1/us]:8.14e+03 at an offset of 126.675 usec -> 2023-04-04T00:25:59.657.066.400 (XO: 27.657068797)
   Min dTM/dt [1/us]:-5.76e+03 at an offset of 126.238 usec -> 2023-04-04T00:25:59.657.065.962 (XO: 27.657068359)

 - Block number: 3 starting at 2023-04-04T00:26:12.866.445.725 (XO: 40.866448818 - 40.866754819, t=34.0 degC)  Rank: 00068
   TM values: Max =   180  Min =  -220  Ave =    -4.45  Std =    17.62
   Min TM value:-220 at an offset of 133.988 usec -> 2023-04-04T00:26:12.866.579.713 (XO: 40.866582806)
   Max TM value:180 at an offset of 119.238 usec -> 2023-04-04T00:26:12.866.564.962 (XO: 40.866568055)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 134.101 usec -> 2023-04-04T00:26:12.866.579.825 (XO: 40.866582918)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 119.338 usec -> 2023-04-04T00:26:12.866.565.062 (XO: 40.866568155)

 - Block number: 4 starting at 2023-04-04T00:26:24.883.915.725 (XO: 52.883918872 - 52.884224873, t=34.0 degC)  Rank: 00081
   TM values: Max =   236  Min =  -248  Ave =    -4.42  Std =    36.21
   Min TM value:-248 at an offset of 122.113 usec -> 2023-04-04T00:26:24.884.037.837 (XO: 52.884040985)
   Max TM value:236 at an offset of 112.500 usec -> 2023-04-04T00:26:24.884.028.225 (XO: 52.884031373)
   Max dTM/dt [1/us]:3.20e+03 at an offset of 164.351 usec -> 2023-04-04T00:26:24.884.080.075 (XO: 52.884083223)
   Min dTM/dt [1/us]:-2.65e+03 at an offset of 142.726 usec -> 2023-04-04T00:26:24.884.058.450 (XO: 52.884061598)

 - Block number: 5 starting at 2023-04-04T00:26:44.699.495.725 (XO: 72.699498983 - 72.699804984, t=34.0 degC)  Rank: 00068
   TM values: Max =   201  Min =  -224  Ave =    -4.39  Std =    17.42
   Min TM value:-224 at an offset of 128.375 usec -> 2023-04-04T00:26:44.699.624.100 (XO: 72.699627359)
   Max TM value:201 at an offset of 128.100 usec -> 2023-04-04T00:26:44.699.623.825 (XO: 72.699627084)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 128.038 usec -> 2023-04-04T00:26:44.699.623.762 (XO: 72.699627021)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 128.200 usec -> 2023-04-04T00:26:44.699.623.925 (XO: 72.699627184)

 - Block number: 6 starting at 2023-04-04T00:26:55.997.106.725 (XO: 83.997110614 - 83.997416615, t=34.0 degC)  Rank: 00068
   TM values: Max =   169  Min =  -214  Ave =    -4.54  Std =    17.40
   Min TM value:-214 at an offset of 133.938 usec -> 2023-04-04T00:26:55.997.240.663 (XO: 83.997244552)
   Max TM value:169 at an offset of 119.525 usec -> 2023-04-04T00:26:55.997.226.250 (XO: 83.997230139)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 134.076 usec -> 2023-04-04T00:26:55.997.240.800 (XO: 83.997244689)
   Min dTM/dt [1/us]:-1.70e+03 at an offset of 131.301 usec -> 2023-04-04T00:26:55.997.238.025 (XO: 83.997241914)
