-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar 21 20:47:28 2023 UTC
-- File 20221015_130544_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.640083 to 19.019917
-each separately from 3s to 19s and from 19628us to 640372us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.379256 in 1 packets: 18
MIU packet time from 2022-10-15T13:05:50.640.083.000 to 2022-10-15T13:06:06.019.917.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.379256 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.5998581 to 20.5998002
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999421 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20221015_130544_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.2399417 to 5.2402307
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.3791990  -- OK for SL --
   MIU time from 2022-10-15T13:05:50.640.083.000 to 2022-10-15T13:05:50.640.372.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.379256

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.6194297 to 20.6197187
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-10-15T13:06:06.019.628.000 to 2022-10-15T13:06:06.019.917.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.239941700 corrected by 40bit 10MHz counter of 4.599858100 obtained at 1Hz counter of 3. Result: 3.640085968
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-10-15T13:05:50.639.942.725
   Max 1st sample MIU time among all SL packets: 2022-10-15T13:06:06.019.776.725
   Min 1st sample XO time among all SL packets: 3.639945693
   Max 1st sample XO time among all SL packets: 19.019778296
   TM values from -225 to 199 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-10-15T13:05:50.639.942.725 (XO: 3.639945693 - 3.640251694, t=37.2 degC)  Rank: 00081
   TM values: Max =   199  Min =  -224  Ave =    -4.39  Std =    21.53
   Min TM value:-224 at an offset of 154.401 usec -> 2022-10-15T13:05:50.640.097.125 (XO: 3.640100093)
   Max TM value:199 at an offset of 112.625 usec -> 2022-10-15T13:05:50.640.055.350 (XO: 3.640058318)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 110.625 usec -> 2022-10-15T13:05:50.640.053.350 (XO: 3.640056318)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 112.725 usec -> 2022-10-15T13:05:50.640.055.450 (XO: 3.640058418)

 - Block number: 2 starting at 2022-10-15T13:06:06.019.487.725 (XO: 19.019489295 - 19.019795296, t=37.2 degC)  Rank: 00076
   TM values: Max =   197  Min =  -225  Ave =    -4.35  Std =    21.90
   Min TM value:-225 at an offset of 123.338 usec -> 2022-10-15T13:06:06.019.611.062 (XO: 19.019612633)
   Max TM value:197 at an offset of 96.388 usec -> 2022-10-15T13:06:06.019.584.112 (XO: 19.019585683)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 99.063 usec -> 2022-10-15T13:06:06.019.586.787 (XO: 19.019588358)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 96.500 usec -> 2022-10-15T13:06:06.019.584.225 (XO: 19.019585796)
