-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Mar 20 20:05:09 2023 UTC
-- File 20221015_111648_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.481742 to 70.272681
-each separately from 14s to 70s and from 261745us to 482031us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      0.779714 in 1 packets: 18
      25.100442 in 1 packets: 36
      12.099546 in 1 packets: 54
      17.809792 in 1 packets: 72
MIU packet time from 2022-10-15T11:17:05.481.742.000 to 2022-10-15T11:18:01.272.681.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      0.779714 in 1 packets: 18
      25.100442 in 1 packets: 36
      12.099546 in 1 packets: 54
      17.809792 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.5479781 to 71.5477757
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      0.9999965 in 1 packets: 18
      24.9999094 in 1 packets: 36
      11.9999567 in 1 packets: 54
      17.9999350 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      04 -> 49.8 degC in 54 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
      2 in 36 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
      2 in 36 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20221015_111648_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.0297205 to 16.0300095
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 0.7797110  -- OK for SL --
   MIU time from 2022-10-15T11:17:05.481.742.000 to 2022-10-15T11:17:05.482.031.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 0.779714

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.8097205 to 16.8100095
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.1003510  -- OK for SL --
   MIU time from 2022-10-15T11:17:06.261.745.000 to 2022-10-15T11:17:06.262.034.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.100442

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.9103605 to 41.9106495
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.0995030  -- OK for SL --
   MIU time from 2022-10-15T11:17:31.362.476.000 to 2022-10-15T11:17:31.362.765.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.099546

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 54.0101525 to 54.0104415
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.8097270  -- OK for SL --
   MIU time from 2022-10-15T11:17:43.462.311.000 to 2022-10-15T11:17:43.462.600.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.809792

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 71.8201685 to 71.8204575
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-10-15T11:18:01.272.392.000 to 2022-10-15T11:18:01.272.681.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.029720500 corrected by 40bit 10MHz counter of 15.547978100 obtained at 1Hz counter of 14. Result: 14.481744086
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-10-15T11:17:05.481.601.725
   Max 1st sample MIU time among all SL packets: 2022-10-15T11:18:01.272.540.725
   Min 1st sample XO time among all SL packets: 14.481603811
   Max 1st sample XO time among all SL packets: 70.272542533
   TM values from -226 to 204 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-10-15T11:17:05.481.601.725 (XO: 14.481603811 - 14.481909812, t=37.2 degC)  Rank: 00080
   TM values: Max =   195  Min =  -221  Ave =    -4.39  Std =    20.95
   Min TM value:-221 at an offset of 106.700 usec -> 2022-10-15T11:17:05.481.708.425 (XO: 14.481710511)
   Max TM value:195 at an offset of 106.375 usec -> 2022-10-15T11:17:05.481.708.100 (XO: 14.481710186)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 109.138 usec -> 2022-10-15T11:17:05.481.710.862 (XO: 14.481712948)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 106.475 usec -> 2022-10-15T11:17:05.481.708.200 (XO: 14.481710286)

 - Block number: 2 starting at 2022-10-15T11:17:06.261.604.725 (XO: 15.261606593 - 15.261912594, t=37.2 degC)  Rank: 00080
   TM values: Max =   204  Min =  -218  Ave =    -4.36  Std =    16.69
   Min TM value:-218 at an offset of 109.800 usec -> 2022-10-15T11:17:06.261.714.525 (XO: 15.261716393)
   Max TM value:204 at an offset of 109.513 usec -> 2022-10-15T11:17:06.261.714.237 (XO: 15.261716106)
   Max dTM/dt [1/us]:1.47e+03 at an offset of 112.325 usec -> 2022-10-15T11:17:06.261.717.050 (XO: 15.261718918)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 109.625 usec -> 2022-10-15T11:17:06.261.714.350 (XO: 15.261716218)

 - Block number: 3 starting at 2022-10-15T11:17:31.362.335.725 (XO: 40.362337529 - 40.362643531, t=37.2 degC)  Rank: 00086
   TM values: Max =   181  Min =  -226  Ave =    -4.28  Std =    19.84
   Min TM value:-226 at an offset of 108.963 usec -> 2022-10-15T11:17:31.362.444.687 (XO: 40.362446492)
   Max TM value:181 at an offset of 78.813 usec -> 2022-10-15T11:17:31.362.414.537 (XO: 40.362416342)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 109.125 usec -> 2022-10-15T11:17:31.362.444.850 (XO: 40.362446655)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 78.913 usec -> 2022-10-15T11:17:31.362.414.637 (XO: 40.362416442)

 - Block number: 4 starting at 2022-10-15T11:17:43.462.170.725 (XO: 52.462173189 - 52.462479190, t=37.2 degC)  Rank: 00079
   TM values: Max =   191  Min =  -223  Ave =    -4.35  Std =    18.03
   Min TM value:-223 at an offset of 120.113 usec -> 2022-10-15T11:17:43.462.290.837 (XO: 52.462293302)
   Max TM value:191 at an offset of 107.875 usec -> 2022-10-15T11:17:43.462.278.600 (XO: 52.462281064)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 122.613 usec -> 2022-10-15T11:17:43.462.293.337 (XO: 52.462295802)
   Min dTM/dt [1/us]:-1.76e+03 at an offset of 119.888 usec -> 2022-10-15T11:17:43.462.290.612 (XO: 52.462293077)

 - Block number: 5 starting at 2022-10-15T11:18:01.272.251.725 (XO: 70.272253532 - 70.272559533, t=37.2 degC)  Rank: 00076
   TM values: Max =   165  Min =  -226  Ave =    -4.37  Std =    20.84
   Min TM value:-226 at an offset of 120.775 usec -> 2022-10-15T11:18:01.272.372.500 (XO: 70.272374308)
   Max TM value:165 at an offset of 121.050 usec -> 2022-10-15T11:18:01.272.372.775 (XO: 70.272374583)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 120.900 usec -> 2022-10-15T11:18:01.272.372.625 (XO: 70.272374433)
   Min dTM/dt [1/us]:-1.58e+03 at an offset of 118.138 usec -> 2022-10-15T11:18:01.272.369.862 (XO: 70.272371670)
