-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar 21 21:54:45 2023 UTC
-- File 20220711_182554_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177030
-each separately from 0s to 14s and from 0us to 841256us.
   Differences in neigboring packets: 
      0.833940 in 2 packets: 1, 13
      0.833943 in 7 packets: 2, 4, 6, 8, 10, 15, 17
      0.833944 in 7 packets: 3, 5, 9, 11, 12, 14, 16
      0.833941 in 1 packets: 7
MIU packet time from 2022-07-11T18:25:57.000.000.000 to 2022-07-11T18:26:11.177.030.000
   Differences in neigboring packets: 
      0.833940 in 2 packets: 1, 13
      0.833943 in 7 packets: 2, 4, 6, 8, 10, 15, 17
      0.833944 in 7 packets: 3, 5, 9, 11, 12, 14, 16
      0.833941 in 1 packets: 7
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.1527349 to 16.1526836
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999963 in 9 packets: 2, 4, 5, 8, 10, 11, 12, 15, 16
      0.9999964 in 5 packets: 3, 6, 9, 14, 17
   XO PPM: 
      -3.7 in 9 packets: 2, 4, 5, 8, 10, 11, 12, 15, 16
      -3.6 in 5 packets: 3, 6, 9, 14, 17
Temperature FPGA: 
      04 -> 49.8 degC in 11 packets: 1, 3, 6, 7, 8, 9, 10, 12, 15, 16, 18
      05 -> 48.5 degC in 6 packets: 2, 4, 5, 13, 14, 17
      03 -> 51.1 degC in 1 packets: 11
Temperature XO: 
      C1 -> 37.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20220711_182554_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.1527350 to 16.3297150
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-11T18:25:57.000.000.000 to 2022-07-11T18:26:11.177.030.000
      Difference between neigboring packets: 
      0.833940 in 2 packets: 1, 13
      0.833943 in 7 packets: 2, 4, 6, 8, 10, 15, 17
      0.833944 in 7 packets: 3, 5, 9, 11, 12, 14, 16
      0.833941 in 1 packets: 7
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.152735000 corrected by 40bit 10MHz counter of 2.152734900 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SR packets: 2022-07-11T18:25:56.999.999.925
   Max 1st sample MIU time among all SR packets: 2022-07-11T18:26:11.177.029.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177031962
   TM values from -66 to 58 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2022-07-11T18:25:56.999.999.925 (XO: 0.000000025 - 0.000017025, t=37.2 degC)
   TM values: Max =    40  Min =   -45  Ave =    -4.33  Std =    15.66

 - Packet number: 2 starting at 2022-07-11T18:25:57.833.939.925 (XO: 0.833943111 - 0.833960111, t=37.2 degC)
   TM values: Max =    34  Min =   -48  Ave =    -4.46  Std =    13.39

 - Packet number: 3 starting at 2022-07-11T18:25:58.667.882.925 (XO: 1.667886129 - 1.667903129, t=37.2 degC)
   TM values: Max =    50  Min =   -66  Ave =    -4.37  Std =    19.43

 - Packet number: 4 starting at 2022-07-11T18:25:59.501.826.925 (XO: 2.501829182 - 2.501846182, t=37.2 degC)
   TM values: Max =    44  Min =   -49  Ave =    -4.50  Std =    14.41

 - Packet number: 5 starting at 2022-07-11T18:26:00.335.769.925 (XO: 3.335772267 - 3.335789267, t=37.2 degC)
   TM values: Max =    48  Min =   -53  Ave =    -4.26  Std =    12.50

 - Packet number: 6 starting at 2022-07-11T18:26:01.169.713.925 (XO: 4.169715336 - 4.169732336, t=37.2 degC)
   TM values: Max =    41  Min =   -56  Ave =    -4.31  Std =    14.86

 - Packet number: 7 starting at 2022-07-11T18:26:02.003.656.925 (XO: 5.003658339 - 5.003675339, t=37.2 degC)
   TM values: Max =    48  Min =   -60  Ave =    -4.48  Std =    17.33

 - Packet number: 8 starting at 2022-07-11T18:26:02.837.597.925 (XO: 5.837601424 - 5.837618424, t=37.2 degC)
   TM values: Max =    48  Min =   -55  Ave =    -4.32  Std =    15.69

 - Packet number: 9 starting at 2022-07-11T18:26:03.671.540.925 (XO: 6.671544443 - 6.671561443, t=37.2 degC)
   TM values: Max =    47  Min =   -49  Ave =    -4.49  Std =    16.62

 - Packet number: 10 starting at 2022-07-11T18:26:04.505.484.925 (XO: 7.505487495 - 7.505504495, t=37.2 degC)
   TM values: Max =    51  Min =   -49  Ave =    -4.21  Std =    16.77

 - Packet number: 11 starting at 2022-07-11T18:26:05.339.427.925 (XO: 8.339430581 - 8.339447581, t=37.2 degC)
   TM values: Max =    58  Min =   -63  Ave =    -4.43  Std =    17.57

 - Packet number: 12 starting at 2022-07-11T18:26:06.173.371.925 (XO: 9.173373666 - 9.173390667, t=37.2 degC)
   TM values: Max =    37  Min =   -49  Ave =    -4.45  Std =    15.53

 - Packet number: 13 starting at 2022-07-11T18:26:07.007.315.925 (XO: 10.007316751 - 10.007333751, t=37.2 degC)
   TM values: Max =    54  Min =   -56  Ave =    -4.25  Std =    16.92

 - Packet number: 14 starting at 2022-07-11T18:26:07.841.255.925 (XO: 10.841259754 - 10.841276754, t=37.2 degC)
   TM values: Max =    37  Min =   -50  Ave =    -4.38  Std =    15.31

 - Packet number: 15 starting at 2022-07-11T18:26:08.675.199.925 (XO: 11.675202823 - 11.675219823, t=37.2 degC)
   TM values: Max =    45  Min =   -52  Ave =    -4.43  Std =    16.57

 - Packet number: 16 starting at 2022-07-11T18:26:09.509.142.925 (XO: 12.509145909 - 12.509162909, t=37.2 degC)
   TM values: Max =    45  Min =   -54  Ave =    -4.49  Std =    16.86

 - Packet number: 17 starting at 2022-07-11T18:26:10.343.086.925 (XO: 13.343088960 - 13.343105960, t=37.2 degC)
   TM values: Max =    50  Min =   -57  Ave =    -4.43  Std =    15.90

 - Packet number: 18 starting at 2022-07-11T18:26:11.177.029.925 (XO: 14.177031962 - 14.177048962, t=37.2 degC)
   TM values: Max =    45  Min =   -51  Ave =    -4.41  Std =    16.22
