-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 00:02:24 2023 UTC
-- File 20220710_215429_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.080427 to 22.870685
-each separately from 13s to 22s and from 80427us to 870685us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.789680 in 1 packets: 18
MIU packet time from 2022-07-10T21:54:45.080.427.000 to 2022-07-10T21:54:54.870.685.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.789680 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.2591563 to 24.2591237
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999674 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220710_215429_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.3395845 to 15.3398735
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.7896470  -- OK for SL --
   MIU time from 2022-07-10T21:54:45.080.427.000 to 2022-07-10T21:54:45.080.716.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.789680

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.1295205 to 25.1298095
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-10T21:54:54.870.396.000 to 2022-07-10T21:54:54.870.685.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.339584500 corrected by 40bit 10MHz counter of 15.259156300 obtained at 1Hz counter of 13. Result: 13.080428498
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-07-10T21:54:45.080.286.725
   Max 1st sample MIU time among all SL packets: 2022-07-10T21:54:54.870.544.725
   Min 1st sample XO time among all SL packets: 13.080288222
   Max 1st sample XO time among all SL packets: 22.870548746
   TM values from -281 to 240 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-07-10T21:54:45.080.286.725 (XO: 13.080288222 - 13.080594223, t=38.8 degC)  Rank: 00088
   TM values: Max =   240  Min =  -281  Ave =    -4.40  Std =    22.38
   Min TM value:-281 at an offset of 112.200 usec -> 2022-07-10T21:54:45.080.398.925 (XO: 13.080400422)
   Max TM value:240 at an offset of 111.950 usec -> 2022-07-10T21:54:45.080.398.675 (XO: 13.080400172)
   Max dTM/dt [1/us]:2.22e+03 at an offset of 114.563 usec -> 2022-07-10T21:54:45.080.401.287 (XO: 13.080402785)
   Min dTM/dt [1/us]:-2.80e+03 at an offset of 112.075 usec -> 2022-07-10T21:54:45.080.398.800 (XO: 13.080400297)

 - Block number: 2 starting at 2022-07-10T21:54:54.870.255.725 (XO: 22.870259745 - 22.870565746, t=38.8 degC)  Rank: 00086
   TM values: Max =   237  Min =  -249  Ave =    -4.47  Std =    30.76
   Min TM value:-249 at an offset of 150.388 usec -> 2022-07-10T21:54:54.870.406.113 (XO: 22.870410133)
   Max TM value:237 at an offset of 126.225 usec -> 2022-07-10T21:54:54.870.381.950 (XO: 22.870385970)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 138.051 usec -> 2022-07-10T21:54:54.870.393.775 (XO: 22.870397795)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 150.238 usec -> 2022-07-10T21:54:54.870.405.963 (XO: 22.870409983)
