-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 23:57:51 2023 UTC
-- File 20220710_215110_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.219292 to 17.539595
-each separately from 13s to 17s and from 219292us to 539595us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.319725 in 1 packets: 18
MIU packet time from 2022-07-10T21:51:26.219.292.000 to 2022-07-10T21:51:30.539.595.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.319725 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.3066160 to 19.3066016
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.9999856 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220710_215110_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.5259088 to 15.5261978
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.3197110  -- OK for SL --
   MIU time from 2022-07-10T21:51:26.219.292.000 to 2022-07-10T21:51:26.219.581.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.319725

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.8459088 to 19.8461978
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-10T21:51:30.539.306.000 to 2022-07-10T21:51:30.539.595.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.525908800 corrected by 40bit 10MHz counter of 15.306616000 obtained at 1Hz counter of 13. Result: 13.219293589
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-07-10T21:51:26.219.151.725
   Max 1st sample MIU time among all SL packets: 2022-07-10T21:51:30.539.454.725
   Min 1st sample XO time among all SL packets: 13.219153314
   Max 1st sample XO time among all SL packets: 17.539457921
   TM values from -285 to 256 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-07-10T21:51:26.219.151.725 (XO: 13.219153314 - 13.219459315, t=38.8 degC)  Rank: 00083
   TM values: Max =   256  Min =  -285  Ave =    -4.44  Std =    30.30
   Min TM value:-285 at an offset of 127.763 usec -> 2022-07-10T21:51:26.219.279.487 (XO: 13.219281077)
   Max TM value:256 at an offset of 125.338 usec -> 2022-07-10T21:51:26.219.277.062 (XO: 13.219278652)
   Max dTM/dt [1/us]:2.50e+03 at an offset of 123.338 usec -> 2022-07-10T21:51:26.219.275.062 (XO: 13.219276652)
   Min dTM/dt [1/us]:-2.72e+03 at an offset of 125.413 usec -> 2022-07-10T21:51:26.219.277.137 (XO: 13.219278727)

 - Block number: 2 starting at 2022-07-10T21:51:30.539.165.725 (XO: 17.539168920 - 17.539474921, t=38.8 degC)  Rank: 00080
   TM values: Max =   202  Min =  -276  Ave =    -4.44  Std =    24.82
   Min TM value:-276 at an offset of 115.700 usec -> 2022-07-10T21:51:30.539.281.425 (XO: 17.539284620)
   Max TM value:202 at an offset of 113.200 usec -> 2022-07-10T21:51:30.539.278.925 (XO: 17.539282120)
   Max dTM/dt [1/us]:2.78e+03 at an offset of 115.800 usec -> 2022-07-10T21:51:30.539.281.525 (XO: 17.539284720)
   Min dTM/dt [1/us]:-2.31e+03 at an offset of 113.325 usec -> 2022-07-10T21:51:30.539.279.050 (XO: 17.539282245)
