-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 24 16:54:56 2023 UTC
-- File 20220709_170255_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.170092 to 29.790679
-each separately from 13s to 29s and from 170092us to 790679us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.620009 in 1 packets: 18
MIU packet time from 2022-07-09T17:03:11.170.092.000 to 2022-07-09T17:03:27.790.679.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.620009 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.7125991 to 30.7125410
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999419 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220709_170255_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.8826918 to 14.8829808
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.6199510  -- OK for SL --
   MIU time from 2022-07-09T17:03:11.170.092.000 to 2022-07-09T17:03:11.170.381.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.620009

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.5029318 to 31.5032208
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-09T17:03:27.790.390.000 to 2022-07-09T17:03:27.790.679.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 14.882691800 corrected by 40bit 10MHz counter of 14.712599100 obtained at 1Hz counter of 13. Result: 13.170093312
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-07-09T17:03:11.169.951.725
   Max 1st sample MIU time among all SL packets: 2022-07-09T17:03:27.790.538.725
   Min 1st sample XO time among all SL packets: 13.169953037
   Max 1st sample XO time among all SL packets: 29.790542371
   TM values from -340 to 285 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-07-09T17:03:11.169.951.725 (XO: 13.169953037 - 13.170259038, t=38.8 degC)  Rank: 00079
   TM values: Max =   285  Min =  -340  Ave =    -4.57  Std =    42.69
   Min TM value:-340 at an offset of 130.238 usec -> 2022-07-09T17:03:11.170.081.962 (XO: 13.170083275)
   Max TM value:285 at an offset of 132.638 usec -> 2022-07-09T17:03:11.170.084.362 (XO: 13.170085675)
   Max dTM/dt [1/us]:3.10e+03 at an offset of 132.513 usec -> 2022-07-09T17:03:11.170.084.237 (XO: 13.170085550)
   Min dTM/dt [1/us]:-2.82e+03 at an offset of 130.038 usec -> 2022-07-09T17:03:11.170.081.762 (XO: 13.170083075)

 - Block number: 2 starting at 2022-07-09T17:03:27.790.249.725 (XO: 29.790253370 - 29.790559371, t=38.8 degC)  Rank: 00080
   TM values: Max =   216  Min =  -300  Ave =    -4.46  Std =    32.06
   Min TM value:-300 at an offset of 129.113 usec -> 2022-07-09T17:03:27.790.378.837 (XO: 29.790382483)
   Max TM value:216 at an offset of 126.638 usec -> 2022-07-09T17:03:27.790.376.362 (XO: 29.790380008)
   Max dTM/dt [1/us]:2.90e+03 at an offset of 129.175 usec -> 2022-07-09T17:03:27.790.378.900 (XO: 29.790382545)
   Min dTM/dt [1/us]:-2.48e+03 at an offset of 126.763 usec -> 2022-07-09T17:03:27.790.376.487 (XO: 29.790380133)
