-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 15:22:56 2023 UTC
-- File 20220704_205409_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.886589 to 29.405047
-each separately from 9s to 29s and from 404758us to 886878us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.517880 in 1 packets: 18
MIU packet time from 2022-07-04T20:54:21.886.589.000 to 2022-07-04T20:54:41.405.047.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.517880 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.7625631 to 30.7624908
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999277 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220704_205409_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.6491528 to 11.6494418
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.5178070  -- OK for SL --
   MIU time from 2022-07-04T20:54:21.886.589.000 to 2022-07-04T20:54:21.886.878.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.517880

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.1672488 to 31.1675378
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-04T20:54:41.404.758.000 to 2022-07-04T20:54:41.405.047.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.649152800 corrected by 40bit 10MHz counter of 10.762563100 obtained at 1Hz counter of 9. Result: 9.886592892
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-07-04T20:54:21.886.448.725
   Max 1st sample MIU time among all SL packets: 2022-07-04T20:54:41.404.906.725
   Min 1st sample XO time among all SL packets: 9.886452616
   Max 1st sample XO time among all SL packets: 29.404908183
   TM values from -308 to 250 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-07-04T20:54:21.886.448.725 (XO: 9.886452616 - 9.886758617, t=38.8 degC)  Rank: 00085
   TM values: Max =   250  Min =  -297  Ave =    -4.44  Std =    36.25
   Min TM value:-297 at an offset of 196.001 usec -> 2022-07-04T20:54:21.886.644.725 (XO: 9.886648617)
   Max TM value:250 at an offset of 122.863 usec -> 2022-07-04T20:54:21.886.571.587 (XO: 9.886575479)
   Max dTM/dt [1/us]:2.79e+03 at an offset of 125.450 usec -> 2022-07-04T20:54:21.886.574.175 (XO: 9.886578067)
   Min dTM/dt [1/us]:-2.68e+03 at an offset of 123.000 usec -> 2022-07-04T20:54:21.886.571.725 (XO: 9.886575617)

 - Block number: 2 starting at 2022-07-04T20:54:41.404.617.725 (XO: 29.404619182 - 29.404925183, t=38.8 degC)  Rank: 00079
   TM values: Max =   249  Min =  -308  Ave =    -4.47  Std =    38.03
   Min TM value:-308 at an offset of 110.588 usec -> 2022-07-04T20:54:41.404.728.312 (XO: 29.404729770)
   Max TM value:249 at an offset of 108.588 usec -> 2022-07-04T20:54:41.404.726.312 (XO: 29.404727770)
   Max dTM/dt [1/us]:2.57e+03 at an offset of 112.913 usec -> 2022-07-04T20:54:41.404.730.637 (XO: 29.404732095)
   Min dTM/dt [1/us]:-3.08e+03 at an offset of 110.475 usec -> 2022-07-04T20:54:41.404.728.200 (XO: 29.404729657)
