-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 13:23:42 2023 UTC
-- File 20220704_183553_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177029
-each separately from 0s to 14s and from 0us to 841255us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 8 packets: 2, 4, 6, 8, 10, 12, 14, 16
      0.833943 in 6 packets: 3, 5, 9, 11, 15, 17
      0.833940 in 2 packets: 7, 13
MIU packet time from 2022-07-04T18:35:56.000.000.000 to 2022-07-04T18:36:10.177.029.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 8 packets: 2, 4, 6, 8, 10, 12, 14, 16
      0.833943 in 6 packets: 3, 5, 9, 11, 15, 17
      0.833940 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.1121386 to 16.1120882
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999964 in 14 packets: 2, 3, 4, 5, 6, 8, 9, 10, 11, 12, 14, 15, 16, 17
   XO PPM: 
      -3.6 in 14 packets (all packets)
Temperature FPGA: 
      04 -> 49.8 degC in 6 packets: 1, 7, 8, 13, 14, 17
      02 -> 52.4 degC in 4 packets: 2, 12, 16, 18
      03 -> 51.1 degC in 8 packets: 3, 4, 5, 6, 9, 10, 11, 15
Temperature XO: 
      C2 -> 38.8 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 13 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13
      1 in 5 packets: 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 13 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13
      1 in 5 packets: 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20220704_183553_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.1121387 to 16.2891187
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-04T18:35:56.000.000.000 to 2022-07-04T18:36:10.177.029.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 8 packets: 2, 4, 6, 8, 10, 12, 14, 16
      0.833943 in 6 packets: 3, 5, 9, 11, 15, 17
      0.833940 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.112138700 corrected by 40bit 10MHz counter of 2.112138600 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SR packets: 2022-07-04T18:35:55.999.999.925
   Max 1st sample MIU time among all SR packets: 2022-07-04T18:36:10.177.028.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177031062
   TM values from -260 to 186 median:-5.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2022-07-04T18:35:55.999.999.925 (XO: 0.000000025 - 0.000017025, t=38.8 degC)
   TM values: Max =    68  Min =   -84  Ave =    -4.42  Std =    27.50

 - Packet number: 2 starting at 2022-07-04T18:35:56.833.938.925 (XO: 0.833943027 - 0.833960027, t=38.8 degC)
   TM values: Max =    84  Min =   -90  Ave =    -4.65  Std =    36.18

 - Packet number: 3 starting at 2022-07-04T18:35:57.667.882.925 (XO: 1.667886029 - 1.667903029, t=38.8 degC)
   TM values: Max =    94  Min =  -106  Ave =    -4.45  Std =    47.46

 - Packet number: 4 starting at 2022-07-04T18:35:58.501.825.925 (XO: 2.501829032 - 2.501846032, t=38.8 degC)
   TM values: Max =    98  Min =  -105  Ave =    -4.48  Std =    51.87

 - Packet number: 5 starting at 2022-07-04T18:35:59.335.769.925 (XO: 3.335772034 - 3.335789034, t=38.8 degC)
   TM values: Max =    85  Min =  -103  Ave =    -4.41  Std =    39.56

 - Packet number: 6 starting at 2022-07-04T18:36:00.169.712.925 (XO: 4.169715036 - 4.169732036, t=38.8 degC)
   TM values: Max =   110  Min =  -124  Ave =    -4.53  Std =    47.36

 - Packet number: 7 starting at 2022-07-04T18:36:01.003.656.925 (XO: 5.003658038 - 5.003675038, t=38.8 degC)
   TM values: Max =   101  Min =  -108  Ave =    -4.42  Std =    47.05

 - Packet number: 8 starting at 2022-07-04T18:36:01.837.596.925 (XO: 5.837601040 - 5.837618040, t=38.8 degC)
   TM values: Max =    72  Min =   -80  Ave =    -4.34  Std =    28.23

 - Packet number: 9 starting at 2022-07-04T18:36:02.671.540.925 (XO: 6.671544043 - 6.671561043, t=38.8 degC)
   TM values: Max =    57  Min =   -63  Ave =    -4.62  Std =    19.77

 - Packet number: 10 starting at 2022-07-04T18:36:03.505.483.925 (XO: 7.505487045 - 7.505504045, t=38.8 degC)
   TM values: Max =    62  Min =   -64  Ave =    -4.42  Std =    23.81

 - Packet number: 11 starting at 2022-07-04T18:36:04.339.427.925 (XO: 8.339430047 - 8.339447047, t=38.8 degC)
   TM values: Max =    61  Min =   -82  Ave =    -4.56  Std =    21.90

 - Packet number: 12 starting at 2022-07-04T18:36:05.173.370.925 (XO: 9.173373049 - 9.173390049, t=38.8 degC)
   TM values: Max =   186  Min =  -260  Ave =    -5.71  Std =    52.87

 - Packet number: 13 starting at 2022-07-04T18:36:06.007.314.925 (XO: 10.007316051 - 10.007333051, t=38.8 degC)
   TM values: Max =    46  Min =   -60  Ave =    -4.42  Std =    16.32

 - Packet number: 14 starting at 2022-07-04T18:36:06.841.254.925 (XO: 10.841259054 - 10.841276054, t=38.8 degC)
   TM values: Max =    87  Min =   -84  Ave =    -4.39  Std =    25.25

 - Packet number: 15 starting at 2022-07-04T18:36:07.675.198.925 (XO: 11.675202056 - 11.675219056, t=38.8 degC)
   TM values: Max =    88  Min =   -99  Ave =    -4.73  Std =    40.65

 - Packet number: 16 starting at 2022-07-04T18:36:08.509.141.925 (XO: 12.509145058 - 12.509162058, t=38.8 degC)
   TM values: Max =    79  Min =   -87  Ave =    -4.30  Std =    36.85

 - Packet number: 17 starting at 2022-07-04T18:36:09.343.085.925 (XO: 13.343088060 - 13.343105060, t=38.8 degC)
   TM values: Max =   105  Min =  -120  Ave =    -4.46  Std =    44.98

 - Packet number: 18 starting at 2022-07-04T18:36:10.177.028.925 (XO: 14.177031062 - 14.177048062, t=38.8 degC)
   TM values: Max =    81  Min =   -98  Ave =    -4.57  Std =    38.57
