-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 24 18:43:45 2023 UTC
-- File 20220701_233123_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.620935 to 69.074647
-each separately from 3s to 69s and from 74358us to 717861us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.897199 in 1 packets: 18
      8.198860 in 1 packets: 36
      21.537631 in 1 packets: 54
      13.818577 in 1 packets: 72
MIU packet time from 2022-07-01T23:31:29.620.935.000 to 2022-07-01T23:32:35.074.647.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.897199 in 1 packets: 18
      8.198860 in 1 packets: 36
      21.537631 in 1 packets: 54
      13.818577 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.6508099 to 70.6505703
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999202 in 2 packets: 18, 54
      7.9999709 in 1 packets: 36
      13.9999491 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 72 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C2 -> 38.8 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 72 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 72 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220701_233123_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.2717459 to 5.2720349
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.8971190  -- OK for SL --
   MIU time from 2022-07-01T23:31:29.620.935.000 to 2022-07-01T23:31:29.621.224.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.897199

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.1691539 to 27.1694429
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.1988310  -- OK for SL --
   MIU time from 2022-07-01T23:31:51.518.423.000 to 2022-07-01T23:31:51.518.712.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.198860

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.3682739 to 35.3685629
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.5375510  -- OK for SL --
   MIU time from 2022-07-01T23:31:59.717.572.000 to 2022-07-01T23:31:59.717.861.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.537631

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.9061139 to 56.9064029
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.8185270  -- OK for SL --
   MIU time from 2022-07-01T23:32:21.255.492.000 to 2022-07-01T23:32:21.255.781.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.818577

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 70.7249299 to 70.7252189
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-01T23:32:35.074.358.000 to 2022-07-01T23:32:35.074.647.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.271745900 corrected by 40bit 10MHz counter of 4.650809900 obtained at 1Hz counter of 3. Result: 3.620938235
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-07-01T23:31:29.620.794.725
   Max 1st sample MIU time among all SL packets: 2022-07-01T23:32:35.074.506.725
   Min 1st sample XO time among all SL packets: 3.620797960
   Max 1st sample XO time among all SL packets: 69.074508593
   TM values from -248 to 197 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-07-01T23:31:29.620.794.725 (XO: 3.620797960 - 3.621103961, t=38.8 degC)  Rank: 00084
   TM values: Max =   197  Min =  -248  Ave =    -4.30  Std =    22.68
   Min TM value:-248 at an offset of 120.025 usec -> 2022-07-01T23:31:29.620.914.750 (XO: 3.620917985)
   Max TM value:197 at an offset of 117.600 usec -> 2022-07-01T23:31:29.620.912.325 (XO: 3.620915560)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 120.150 usec -> 2022-07-01T23:31:29.620.914.875 (XO: 3.620918110)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 117.713 usec -> 2022-07-01T23:31:29.620.912.437 (XO: 3.620915673)

 - Block number: 2 starting at 2022-07-01T23:31:51.518.282.725 (XO: 25.518285443 - 25.518591444, t=38.8 degC)  Rank: 00076
   TM values: Max =   174  Min =  -221  Ave =    -4.32  Std =    22.37
   Min TM value:-221 at an offset of 119.388 usec -> 2022-07-01T23:31:51.518.402.112 (XO: 25.518404831)
   Max TM value:174 at an offset of 75.188 usec -> 2022-07-01T23:31:51.518.357.912 (XO: 25.518360630)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 119.513 usec -> 2022-07-01T23:31:51.518.402.237 (XO: 25.518404956)
   Min dTM/dt [1/us]:-1.76e+03 at an offset of 75.300 usec -> 2022-07-01T23:31:51.518.358.025 (XO: 25.518360743)

 - Block number: 3 starting at 2022-07-01T23:31:59.717.431.725 (XO: 33.717435208 - 33.717741209, t=38.8 degC)  Rank: 00074
   TM values: Max =   180  Min =  -216  Ave =    -4.34  Std =    23.03
   Min TM value:-216 at an offset of 119.750 usec -> 2022-07-01T23:31:59.717.551.475 (XO: 33.717554958)
   Max TM value:180 at an offset of 120.000 usec -> 2022-07-01T23:31:59.717.551.725 (XO: 33.717555208)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 119.875 usec -> 2022-07-01T23:31:59.717.551.600 (XO: 33.717555083)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 119.663 usec -> 2022-07-01T23:31:59.717.551.387 (XO: 33.717554871)

 - Block number: 4 starting at 2022-07-01T23:32:21.255.351.725 (XO: 55.255353370 - 55.255659371, t=38.8 degC)  Rank: 00078
   TM values: Max =   181  Min =  -222  Ave =    -4.33  Std =    24.49
   Min TM value:-222 at an offset of 120.563 usec -> 2022-07-01T23:32:21.255.472.287 (XO: 55.255473933)
   Max TM value:181 at an offset of 162.013 usec -> 2022-07-01T23:32:21.255.513.738 (XO: 55.255515383)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 120.688 usec -> 2022-07-01T23:32:21.255.472.412 (XO: 55.255474058)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 118.213 usec -> 2022-07-01T23:32:21.255.469.937 (XO: 55.255471583)

 - Block number: 5 starting at 2022-07-01T23:32:35.074.217.725 (XO: 69.074219592 - 69.074525593, t=38.8 degC)  Rank: 00081
   TM values: Max =   175  Min =  -242  Ave =    -4.44  Std =    22.04
   Min TM value:-242 at an offset of 119.263 usec -> 2022-07-01T23:32:35.074.336.987 (XO: 69.074338855)
   Max TM value:175 at an offset of 119.513 usec -> 2022-07-01T23:32:35.074.337.237 (XO: 69.074339105)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 119.388 usec -> 2022-07-01T23:32:35.074.337.112 (XO: 69.074338980)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 116.913 usec -> 2022-07-01T23:32:35.074.334.637 (XO: 69.074336505)
