-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 24 16:32:40 2023 UTC
-- File 20220701_145955_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.661059 to 27.491567
-each separately from 11s to 27s and from 491278us to 661348us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.829930 in 1 packets: 18
MIU packet time from 2022-07-01T15:00:09.661.059.000 to 2022-07-01T15:00:25.491.567.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.829930 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.2643102 to 29.2642516
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999414 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220701_145955_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.9253700 to 13.9256590
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.8298710  -- OK for SL --
   MIU time from 2022-07-01T15:00:09.661.059.000 to 2022-07-01T15:00:09.661.348.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.829930

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      91 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.7555300 to 29.7558190
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-01T15:00:25.491.278.000 to 2022-07-01T15:00:25.491.567.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.925370000 corrected by 40bit 10MHz counter of 13.264310200 obtained at 1Hz counter of 11. Result: 11.661062246
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-07-01T15:00:09.660.918.725
   Max 1st sample MIU time among all SL packets: 2022-07-01T15:00:25.491.426.725
   Min 1st sample XO time among all SL packets: 11.660921970
   Max 1st sample XO time among all SL packets: 27.491428943
   TM values from -287 to 242 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-07-01T15:00:09.660.918.725 (XO: 11.660921970 - 11.661227972, t=37.2 degC)  Rank: 00081
   TM values: Max =   242  Min =  -287  Ave =    -4.40  Std =    33.59
   Min TM value:-287 at an offset of 127.075 usec -> 2022-07-01T15:00:09.661.045.800 (XO: 11.661049046)
   Max TM value:242 at an offset of 124.638 usec -> 2022-07-01T15:00:09.661.043.362 (XO: 11.661046608)
   Max dTM/dt [1/us]:2.41e+03 at an offset of 127.225 usec -> 2022-07-01T15:00:09.661.045.950 (XO: 11.661049196)
   Min dTM/dt [1/us]:-2.61e+03 at an offset of 124.738 usec -> 2022-07-01T15:00:09.661.043.462 (XO: 11.661046708)

 - Block number: 2 starting at 2022-07-01T15:00:25.491.137.725 (XO: 27.491139942 - 27.491445943, t=37.2 degC)  Rank: 00091
   TM values: Max =   221  Min =  -262  Ave =    -4.41  Std =    25.40
   Min TM value:-262 at an offset of 102.788 usec -> 2022-07-01T15:00:25.491.240.512 (XO: 27.491242730)
   Max TM value:221 at an offset of 100.750 usec -> 2022-07-01T15:00:25.491.238.475 (XO: 27.491240693)
   Max dTM/dt [1/us]:2.36e+03 at an offset of 105.100 usec -> 2022-07-01T15:00:25.491.242.825 (XO: 27.491245043)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 102.663 usec -> 2022-07-01T15:00:25.491.240.387 (XO: 27.491242605)
