-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 24 14:20:44 2023 UTC
-- File 20220701_124845_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.711851 to 22.731107
-each separately from 7s to 22s and from 711851us to 731107us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.018678 in 1 packets: 18
MIU packet time from 2022-07-01T12:48:55.711.851.000 to 2022-07-01T12:49:10.731.107.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.018678 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.2514531 to 24.2513981
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999450 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220701_124845_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.9633042 to 9.9635932
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0186230  -- OK for SL --
   MIU time from 2022-07-01T12:48:55.711.851.000 to 2022-07-01T12:48:55.712.140.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.018678

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.9822162 to 24.9825052
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-01T12:49:10.730.818.000 to 2022-07-01T12:49:10.731.107.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.963304200 corrected by 40bit 10MHz counter of 9.251453100 obtained at 1Hz counter of 7. Result: 7.711853734
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-07-01T12:48:55.711.710.725
   Max 1st sample MIU time among all SL packets: 2022-07-01T12:49:10.730.966.725
   Min 1st sample XO time among all SL packets: 7.711713458
   Max 1st sample XO time among all SL packets: 22.730969530
   TM values from -278 to 233 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-07-01T12:48:55.711.710.725 (XO: 7.711713458 - 7.712019459, t=37.2 degC)  Rank: 00084
   TM values: Max =   233  Min =  -278  Ave =    -4.48  Std =    22.35
   Min TM value:-278 at an offset of 117.225 usec -> 2022-07-01T12:48:55.711.827.950 (XO: 7.711830684)
   Max TM value:233 at an offset of 114.763 usec -> 2022-07-01T12:48:55.711.825.487 (XO: 7.711828221)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 117.338 usec -> 2022-07-01T12:48:55.711.828.062 (XO: 7.711830796)
   Min dTM/dt [1/us]:-2.64e+03 at an offset of 114.863 usec -> 2022-07-01T12:48:55.711.825.587 (XO: 7.711828321)

 - Block number: 2 starting at 2022-07-01T12:49:10.730.677.725 (XO: 22.730680529 - 22.730986530, t=37.2 degC)  Rank: 00079
   TM values: Max =   227  Min =  -253  Ave =    -4.23  Std =    28.88
   Min TM value:-253 at an offset of 117.363 usec -> 2022-07-01T12:49:10.730.795.087 (XO: 22.730797891)
   Max TM value:227 at an offset of 114.888 usec -> 2022-07-01T12:49:10.730.792.612 (XO: 22.730795416)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 117.513 usec -> 2022-07-01T12:49:10.730.795.237 (XO: 22.730798041)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 114.988 usec -> 2022-07-01T12:49:10.730.792.712 (XO: 22.730795516)
