-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 24 11:25:09 2023 UTC
-- File 20220701_112031_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.466493 to 24.666853
-each separately from 9s to 24s and from 466493us to 666853us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.199782 in 1 packets: 18
MIU packet time from 2022-07-01T11:20:43.466.493.000 to 2022-07-01T11:20:58.666.853.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.199782 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.1127004 to 26.1126461
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999457 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220701_112031_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.5791948 to 11.5794838
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.1997270  -- OK for SL --
   MIU time from 2022-07-01T11:20:43.466.493.000 to 2022-07-01T11:20:43.466.782.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.199782

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.7792108 to 26.7794998
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-01T11:20:58.666.564.000 to 2022-07-01T11:20:58.666.853.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.579194800 corrected by 40bit 10MHz counter of 11.112700400 obtained at 1Hz counter of 9. Result: 9.466496079
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-07-01T11:20:43.466.352.725
   Max 1st sample MIU time among all SL packets: 2022-07-01T11:20:58.666.712.725
   Min 1st sample XO time among all SL packets: 9.466355804
   Max 1st sample XO time among all SL packets: 24.666715825
   TM values from -261 to 243 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-07-01T11:20:43.466.352.725 (XO: 9.466355804 - 9.466661805, t=38.8 degC)  Rank: 00086
   TM values: Max =   243  Min =  -261  Ave =    -4.40  Std =    26.50
   Min TM value:-261 at an offset of 122.200 usec -> 2022-07-01T11:20:43.466.474.925 (XO: 9.466478004)
   Max TM value:243 at an offset of 119.763 usec -> 2022-07-01T11:20:43.466.472.487 (XO: 9.466475567)
   Max dTM/dt [1/us]:2.28e+03 at an offset of 122.313 usec -> 2022-07-01T11:20:43.466.475.037 (XO: 9.466478117)
   Min dTM/dt [1/us]:-2.44e+03 at an offset of 119.863 usec -> 2022-07-01T11:20:43.466.472.587 (XO: 9.466475667)

 - Block number: 2 starting at 2022-07-01T11:20:58.666.423.725 (XO: 24.666426824 - 24.666732825, t=38.8 degC)  Rank: 00083
   TM values: Max =   222  Min =  -249  Ave =    -4.47  Std =    27.47
   Min TM value:-249 at an offset of 122.638 usec -> 2022-07-01T11:20:58.666.546.362 (XO: 24.666549462)
   Max TM value:222 at an offset of 120.225 usec -> 2022-07-01T11:20:58.666.543.950 (XO: 24.666547050)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 151.176 usec -> 2022-07-01T11:20:58.666.574.900 (XO: 24.666578000)
   Min dTM/dt [1/us]:-2.39e+03 at an offset of 120.338 usec -> 2022-07-01T11:20:58.666.544.062 (XO: 24.666547162)
