-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 24 10:46:13 2023 UTC
-- File 20220701_104201_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.557120 to 58.557187
-each separately from 9s to 58s and from 516813us to 997291us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.439593 in 1 packets: 18
      22.519522 in 1 packets: 36
      19.039796 in 1 packets: 54
MIU packet time from 2022-07-01T10:42:13.557.120.000 to 2022-07-01T10:43:02.557.187.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.439593 in 1 packets: 18
      22.519522 in 1 packets: 36
      19.039796 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.7597848 to 59.7596077
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.9999747 in 1 packets: 18
      22.9999167 in 1 packets: 36
      18.9999315 in 1 packets: 54
Temperature FPGA: 
      03 -> 51.1 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C2 -> 38.8 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20220701_104201_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.3169062 to 11.3171952
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.4395670  -- OK for SL --
   MIU time from 2022-07-01T10:42:13.557.120.000 to 2022-07-01T10:42:13.557.409.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.439593

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.7567622 to 18.7570512
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.5194390  -- OK for SL --
   MIU time from 2022-07-01T10:42:20.997.002.000 to 2022-07-01T10:42:20.997.291.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.519522

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.2764902 to 41.2767792
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.0397270  -- OK for SL --
   MIU time from 2022-07-01T10:42:43.516.813.000 to 2022-07-01T10:42:43.517.102.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.039796

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 60.3165062 to 60.3167952
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-01T10:43:02.556.898.000 to 2022-07-01T10:43:02.557.187.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.316906200 corrected by 40bit 10MHz counter of 10.759784800 obtained at 1Hz counter of 9. Result: 9.557123406
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-07-01T10:42:13.556.979.725
   Max 1st sample MIU time among all SL packets: 2022-07-01T10:43:02.557.046.725
   Min 1st sample XO time among all SL packets: 9.556983130
   Max 1st sample XO time among all SL packets: 58.557049230
   TM values from -254 to 219 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-07-01T10:42:13.556.979.725 (XO: 9.556983130 - 9.557289131, t=38.8 degC)  Rank: 00082
   TM values: Max =   187  Min =  -254  Ave =    -4.50  Std =    23.75
   Min TM value:-254 at an offset of 125.025 usec -> 2022-07-01T10:42:13.557.104.750 (XO: 9.557108156)
   Max TM value:187 at an offset of 83.150 usec -> 2022-07-01T10:42:13.557.062.875 (XO: 9.557066280)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 127.263 usec -> 2022-07-01T10:42:13.557.106.987 (XO: 9.557110393)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 83.275 usec -> 2022-07-01T10:42:13.557.063.000 (XO: 9.557066405)

 - Block number: 2 starting at 2022-07-01T10:42:20.996.861.725 (XO: 16.996866014 - 16.997172015, t=38.8 degC)  Rank: 00082
   TM values: Max =   214  Min =  -240  Ave =    -4.53  Std =    20.31
   Min TM value:-240 at an offset of 126.000 usec -> 2022-07-01T10:42:20.996.987.725 (XO: 16.996992014)
   Max TM value:214 at an offset of 125.738 usec -> 2022-07-01T10:42:20.996.987.462 (XO: 16.996991752)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 125.663 usec -> 2022-07-01T10:42:20.996.987.387 (XO: 16.996991677)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 125.838 usec -> 2022-07-01T10:42:20.996.987.562 (XO: 16.996991852)

 - Block number: 3 starting at 2022-07-01T10:42:43.516.672.725 (XO: 39.516675585 - 39.516981586, t=38.8 degC)  Rank: 00076
   TM values: Max =   194  Min =  -252  Ave =    -4.46  Std =    28.91
   Min TM value:-252 at an offset of 125.025 usec -> 2022-07-01T10:42:43.516.797.750 (XO: 39.516800610)
   Max TM value:194 at an offset of 124.750 usec -> 2022-07-01T10:42:43.516.797.475 (XO: 39.516800335)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 127.275 usec -> 2022-07-01T10:42:43.516.800.000 (XO: 39.516802860)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 127.088 usec -> 2022-07-01T10:42:43.516.799.812 (XO: 39.516802673)

 - Block number: 4 starting at 2022-07-01T10:43:02.556.757.725 (XO: 58.556760229 - 58.557066230, t=38.8 degC)  Rank: 00076
   TM values: Max =   219  Min =  -240  Ave =    -4.46  Std =    23.08
   Min TM value:-240 at an offset of 126.888 usec -> 2022-07-01T10:43:02.556.884.612 (XO: 58.556887117)
   Max TM value:219 at an offset of 126.625 usec -> 2022-07-01T10:43:02.556.884.350 (XO: 58.556886855)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 129.150 usec -> 2022-07-01T10:43:02.556.886.875 (XO: 58.556889380)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 126.725 usec -> 2022-07-01T10:43:02.556.884.450 (XO: 58.556886955)
