-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 24 10:06:16 2023 UTC
-- File 20220701_100340_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.474270 to 41.530177
-each separately from 1s to 41s and from 474270us to 768490us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.293642 in 1 packets: 18
      14.761398 in 1 packets: 36
MIU packet time from 2022-07-01T10:03:44.474.270.000 to 2022-07-01T10:04:24.530.177.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.293642 in 1 packets: 18
      14.761398 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.6546099 to 42.6544645
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999091 in 1 packets: 18
      14.9999455 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C2 -> 38.8 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220701_100340_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.1288810 to 3.1291700
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.2935510  -- OK for SL --
   MIU time from 2022-07-01T10:03:44.474.270.000 to 2022-07-01T10:03:44.474.559.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.293642

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.4227210 to 28.4230100
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7613430  -- OK for SL --
   MIU time from 2022-07-01T10:04:09.768.201.000 to 2022-07-01T10:04:09.768.490.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.761398

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.1843530 to 43.1846420
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-01T10:04:24.529.888.000 to 2022-07-01T10:04:24.530.177.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.128881000 corrected by 40bit 10MHz counter of 2.654609900 obtained at 1Hz counter of 1. Result: 1.474272807
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-07-01T10:03:44.474.129.725
   Max 1st sample MIU time among all SL packets: 2022-07-01T10:04:24.530.036.725
   Min 1st sample XO time among all SL packets: 1.474132532
   Max 1st sample XO time among all SL packets: 41.530039133
   TM values from -226 to 215 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-07-01T10:03:44.474.129.725 (XO: 1.474132532 - 1.474438533, t=38.8 degC)  Rank: 00081
   TM values: Max =   215  Min =  -226  Ave =    -4.52  Std =    22.14
   Min TM value:-226 at an offset of 108.888 usec -> 2022-07-01T10:03:44.474.238.612 (XO: 1.474241420)
   Max TM value:215 at an offset of 108.625 usec -> 2022-07-01T10:03:44.474.238.350 (XO: 1.474241157)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 111.188 usec -> 2022-07-01T10:03:44.474.240.912 (XO: 1.474243720)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 108.725 usec -> 2022-07-01T10:03:44.474.238.450 (XO: 1.474241257)

 - Block number: 2 starting at 2022-07-01T10:04:09.768.060.725 (XO: 26.768064490 - 26.768370491, t=38.8 degC)  Rank: 00077
   TM values: Max =   187  Min =  -221  Ave =    -4.50  Std =    27.32
   Min TM value:-221 at an offset of 117.625 usec -> 2022-07-01T10:04:09.768.178.350 (XO: 26.768182115)
   Max TM value:187 at an offset of 120.050 usec -> 2022-07-01T10:04:09.768.180.775 (XO: 26.768184540)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 119.950 usec -> 2022-07-01T10:04:09.768.180.675 (XO: 26.768184440)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 117.525 usec -> 2022-07-01T10:04:09.768.178.250 (XO: 26.768182015)

 - Block number: 3 starting at 2022-07-01T10:04:24.529.747.725 (XO: 41.529750132 - 41.530056133, t=38.8 degC)  Rank: 00083
   TM values: Max =   194  Min =  -220  Ave =    -4.47  Std =    23.37
   Min TM value:-220 at an offset of 119.263 usec -> 2022-07-01T10:04:24.529.866.987 (XO: 41.529869395)
   Max TM value:194 at an offset of 117.213 usec -> 2022-07-01T10:04:24.529.864.937 (XO: 41.529867345)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 117.100 usec -> 2022-07-01T10:04:24.529.864.825 (XO: 41.529867233)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 119.100 usec -> 2022-07-01T10:04:24.529.866.825 (XO: 41.529869233)
