-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 24 09:48:14 2023 UTC
-- File 20220701_094856_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.904100 to 16.803080
-each separately from 2s to 16s and from 802791us to 904389us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.898402 in 1 packets: 18
MIU packet time from 2022-07-01T09:49:01.904.100.000 to 2022-07-01T09:49:15.803.080.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.898402 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.2479294 to 18.2478789
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999495 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220701_094856_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.1520306 to 5.1523196
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.8983510  -- OK for SL --
   MIU time from 2022-07-01T09:49:01.904.100.000 to 2022-07-01T09:49:01.904.389.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.898402

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.0506706 to 19.0509596
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-01T09:49:15.802.791.000 to 2022-07-01T09:49:15.803.080.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 5.152030600 corrected by 40bit 10MHz counter of 4.247929400 obtained at 1Hz counter of 2. Result: 2.904104455
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-07-01T09:49:01.903.959.725
   Max 1st sample MIU time among all SL packets: 2022-07-01T09:49:15.802.939.725
   Min 1st sample XO time among all SL packets: 2.903964179
   Max 1st sample XO time among all SL packets: 16.802943316
   TM values from -232 to 197 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-07-01T09:49:01.903.959.725 (XO: 2.903964179 - 2.904270180, t=38.8 degC)  Rank: 00081
   TM values: Max =   197  Min =  -232  Ave =    -4.55  Std =    22.76
   Min TM value:-232 at an offset of 109.638 usec -> 2022-07-01T09:49:01.904.069.362 (XO: 2.904073817)
   Max TM value:197 at an offset of 109.388 usec -> 2022-07-01T09:49:01.904.069.112 (XO: 2.904073567)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 107.475 usec -> 2022-07-01T09:49:01.904.067.200 (XO: 2.904071655)
   Min dTM/dt [1/us]:-2.21e+03 at an offset of 109.513 usec -> 2022-07-01T09:49:01.904.069.237 (XO: 2.904073692)

 - Block number: 2 starting at 2022-07-01T09:49:15.802.650.725 (XO: 16.802654315 - 16.802960316, t=38.8 degC)  Rank: 00079
   TM values: Max =   175  Min =  -221  Ave =    -4.53  Std =    19.44
   Min TM value:-221 at an offset of 122.825 usec -> 2022-07-01T09:49:15.802.773.550 (XO: 16.802777140)
   Max TM value:175 at an offset of 120.363 usec -> 2022-07-01T09:49:15.802.771.087 (XO: 16.802774677)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 122.950 usec -> 2022-07-01T09:49:15.802.773.675 (XO: 16.802777265)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 120.488 usec -> 2022-07-01T09:49:15.802.771.212 (XO: 16.802774802)
