-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 16:23:31 2023 UTC
-- File 20220630_162159_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.665665 to 19.366381
-each separately from 7s to 19s and from 366092us to 665954us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.700138 in 1 packets: 18
MIU packet time from 2022-06-30T16:22:09.665.665.000 to 2022-06-30T16:22:21.366.381.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.700138 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.6181232 to 20.6180794
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999562 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220630_162159_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.2837887 to 9.2840777
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.7000950  -- OK for SL --
   MIU time from 2022-06-30T16:22:09.665.665.000 to 2022-06-30T16:22:09.665.954.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.700138

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.9841727 to 20.9844617
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-30T16:22:21.366.092.000 to 2022-06-30T16:22:21.366.381.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.283788700 corrected by 40bit 10MHz counter of 8.618123200 obtained at 1Hz counter of 7. Result: 7.665667963
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-06-30T16:22:09.665.524.725
   Max 1st sample MIU time among all SL packets: 2022-06-30T16:22:21.366.240.725
   Min 1st sample XO time among all SL packets: 7.665527687
   Max 1st sample XO time among all SL packets: 19.366243343
   TM values from -250 to 198 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-30T16:22:09.665.524.725 (XO: 7.665527687 - 7.665833689, t=37.2 degC)  Rank: 00080
   TM values: Max =   197  Min =  -248  Ave =    -4.59  Std =    22.35
   Min TM value:-248 at an offset of 84.125 usec -> 2022-06-30T16:22:09.665.608.850 (XO: 7.665611813)
   Max TM value:197 at an offset of 81.663 usec -> 2022-06-30T16:22:09.665.606.387 (XO: 7.665609350)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 84.238 usec -> 2022-06-30T16:22:09.665.608.962 (XO: 7.665611925)
   Min dTM/dt [1/us]:-2.43e+03 at an offset of 81.788 usec -> 2022-06-30T16:22:09.665.606.512 (XO: 7.665609475)

 - Block number: 2 starting at 2022-06-30T16:22:21.365.951.725 (XO: 19.365954342 - 19.366260344, t=37.2 degC)  Rank: 00078
   TM values: Max =   198  Min =  -250  Ave =    -4.47  Std =    21.43
   Min TM value:-250 at an offset of 123.550 usec -> 2022-06-30T16:22:21.366.075.275 (XO: 19.366077893)
   Max TM value:198 at an offset of 121.088 usec -> 2022-06-30T16:22:21.366.072.812 (XO: 19.366075430)
   Max dTM/dt [1/us]:2.28e+03 at an offset of 123.650 usec -> 2022-06-30T16:22:21.366.075.375 (XO: 19.366077993)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 123.450 usec -> 2022-06-30T16:22:21.366.075.175 (XO: 19.366077793)
