-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 11:56:43 2023 UTC
-- File 20220630_124510_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.800565 to 23.439446
-each separately from 10s to 23s and from 439157us to 800854us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.638303 in 1 packets: 18
MIU packet time from 2022-06-30T12:45:24.800.565.000 to 2022-06-30T12:45:37.439.446.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.638303 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.2574730 to 25.2574254
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999524 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220630_124510_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.0580388 to 13.0583278
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6382550  -- OK for SL --
   MIU time from 2022-06-30T12:45:24.800.565.000 to 2022-06-30T12:45:24.800.854.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.638303

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.6965828 to 25.6968718
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-30T12:45:37.439.157.000 to 2022-06-30T12:45:37.439.446.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 13.058038800 corrected by 40bit 10MHz counter of 12.257473000 obtained at 1Hz counter of 10. Result: 10.800568762
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-30T12:45:24.800.424.725
   Max 1st sample MIU time among all SL packets: 2022-06-30T12:45:37.439.305.725
   Min 1st sample XO time among all SL packets: 10.800428487
   Max 1st sample XO time among all SL packets: 23.439307750
   TM values from -237 to 194 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-30T12:45:24.800.424.725 (XO: 10.800428487 - 10.800734488, t=37.2 degC)  Rank: 00087
   TM values: Max =   194  Min =  -237  Ave =    -4.58  Std =    20.68
   Min TM value:-237 at an offset of 123.463 usec -> 2022-06-30T12:45:24.800.548.187 (XO: 10.800551950)
   Max TM value:194 at an offset of 123.713 usec -> 2022-06-30T12:45:24.800.548.437 (XO: 10.800552200)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 123.613 usec -> 2022-06-30T12:45:24.800.548.337 (XO: 10.800552100)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 123.363 usec -> 2022-06-30T12:45:24.800.548.087 (XO: 10.800551850)

 - Block number: 2 starting at 2022-06-30T12:45:37.439.016.725 (XO: 23.439018749 - 23.439324751, t=37.2 degC)  Rank: 00082
   TM values: Max =   165  Min =  -230  Ave =    -4.49  Std =    19.44
   Min TM value:-230 at an offset of 108.150 usec -> 2022-06-30T12:45:37.439.124.875 (XO: 23.439126900)
   Max TM value:165 at an offset of 113.050 usec -> 2022-06-30T12:45:37.439.129.775 (XO: 23.439131800)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 110.450 usec -> 2022-06-30T12:45:37.439.127.175 (XO: 23.439129200)
   Min dTM/dt [1/us]:-1.73e+03 at an offset of 107.913 usec -> 2022-06-30T12:45:37.439.124.637 (XO: 23.439126662)
