-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 09:58:29 2023 UTC
-- File 20220630_113217_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.222982 to 60.617403
-each separately from 2s to 60s and from 79450us to 617403us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.997724 in 1 packets: 18
      14.858166 in 1 packets: 36
      9.388784 in 1 packets: 54
      13.148302 in 1 packets: 72
MIU packet time from 2022-06-30T11:32:22.222.982.000 to 2022-06-30T11:33:20.617.403.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.997724 in 1 packets: 18
      14.858166 in 1 packets: 36
      9.388784 in 1 packets: 54
      13.148302 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.7660119 to 61.7658005
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999235 in 1 packets: 18
      14.9999452 in 1 packets: 36
      8.9999672 in 1 packets: 54
      12.9999527 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 71 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 71 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220630_113217_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.9889953 to 3.9892843
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.9976470  -- OK for SL --
   MIU time from 2022-06-30T11:32:22.222.982.000 to 2022-06-30T11:32:22.223.271.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.997724

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.9869313 to 24.9872203
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8581110  -- OK for SL --
   MIU time from 2022-06-30T11:32:43.220.995.000 to 2022-06-30T11:32:43.221.284.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.858166

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.8453313 to 39.8456203
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.3887510  -- OK for SL --
   MIU time from 2022-06-30T11:32:58.079.450.000 to 2022-06-30T11:32:58.079.739.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.388784

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.2343713 to 49.2346603
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.1482550  -- OK for SL --
   MIU time from 2022-06-30T11:33:07.468.523.000 to 2022-06-30T11:33:07.468.812.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.148302

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 62.3829153 to 62.3832043
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-30T11:33:20.617.114.000 to 2022-06-30T11:33:20.617.403.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 3.988995300 corrected by 40bit 10MHz counter of 3.766011900 obtained at 1Hz counter of 2. Result: 2.222984203
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-06-30T11:32:22.222.841.725
   Max 1st sample MIU time among all SL packets: 2022-06-30T11:33:20.617.262.725
   Min 1st sample XO time among all SL packets: 2.222843927
   Max 1st sample XO time among all SL packets: 60.617265809
   TM values from -248 to 229 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-30T11:32:22.222.841.725 (XO: 2.222843927 - 2.223149928, t=37.2 degC)  Rank: 00082
   TM values: Max =   162  Min =  -211  Ave =    -4.47  Std =    22.09
   Min TM value:-211 at an offset of 128.038 usec -> 2022-06-30T11:32:22.222.969.762 (XO: 2.222971965)
   Max TM value:162 at an offset of 132.925 usec -> 2022-06-30T11:32:22.222.974.650 (XO: 2.222976853)
   Max dTM/dt [1/us]:1.34e+03 at an offset of 154.813 usec -> 2022-06-30T11:32:22.222.996.538 (XO: 2.222998740)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 157.176 usec -> 2022-06-30T11:32:22.222.998.900 (XO: 2.223001103)

 - Block number: 2 starting at 2022-06-30T11:32:43.220.854.725 (XO: 23.220856442 - 23.221162443, t=37.2 degC)  Rank: 00089
   TM values: Max =   229  Min =  -232  Ave =    -4.52  Std =    18.34
   Min TM value:-232 at an offset of 129.225 usec -> 2022-06-30T11:32:43.220.983.950 (XO: 23.220985668)
   Max TM value:229 at an offset of 128.975 usec -> 2022-06-30T11:32:43.220.983.700 (XO: 23.220985418)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 128.900 usec -> 2022-06-30T11:32:43.220.983.625 (XO: 23.220985343)
   Min dTM/dt [1/us]:-2.42e+03 at an offset of 129.088 usec -> 2022-06-30T11:32:43.220.983.812 (XO: 23.220985530)

 - Block number: 3 starting at 2022-06-30T11:32:58.079.309.725 (XO: 38.079310711 - 38.079616712, t=37.2 degC)  Rank: 00079
   TM values: Max =   168  Min =  -234  Ave =    -4.39  Std =    21.89
   Min TM value:-234 at an offset of 132.575 usec -> 2022-06-30T11:32:58.079.442.300 (XO: 38.079443286)
   Max TM value:168 at an offset of 102.975 usec -> 2022-06-30T11:32:58.079.412.700 (XO: 38.079413686)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 132.688 usec -> 2022-06-30T11:32:58.079.442.412 (XO: 38.079443398)
   Min dTM/dt [1/us]:-1.91e+03 at an offset of 132.500 usec -> 2022-06-30T11:32:58.079.442.225 (XO: 38.079443211)

 - Block number: 4 starting at 2022-06-30T11:33:07.468.382.725 (XO: 47.468384911 - 47.468690912, t=37.2 degC)  Rank: 00079
   TM values: Max =   180  Min =  -222  Ave =    -4.48  Std =    19.26
   Min TM value:-222 at an offset of 130.650 usec -> 2022-06-30T11:33:07.468.513.375 (XO: 47.468515562)
   Max TM value:180 at an offset of 128.200 usec -> 2022-06-30T11:33:07.468.510.925 (XO: 47.468513112)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 130.750 usec -> 2022-06-30T11:33:07.468.513.475 (XO: 47.468515662)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 128.325 usec -> 2022-06-30T11:33:07.468.511.050 (XO: 47.468513237)

 - Block number: 5 starting at 2022-06-30T11:33:20.616.973.725 (XO: 60.616976808 - 60.617282809, t=37.2 degC)  Rank: 00079
   TM values: Max =   194  Min =  -248  Ave =    -4.55  Std =    22.50
   Min TM value:-248 at an offset of 133.263 usec -> 2022-06-30T11:33:20.617.106.987 (XO: 60.617110071)
   Max TM value:194 at an offset of 130.813 usec -> 2022-06-30T11:33:20.617.104.537 (XO: 60.617107621)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 133.363 usec -> 2022-06-30T11:33:20.617.107.087 (XO: 60.617110171)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 133.175 usec -> 2022-06-30T11:33:20.617.106.900 (XO: 60.617109983)
