-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 03:19:05 2023 UTC
-- File 20220629_165456_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.763534 to 29.874491
-each separately from 0s to 29s and from 763534us to 874491us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      29.110379 in 1 packets: 18
MIU packet time from 2022-06-29T16:54:59.763.534.000 to 2022-06-29T16:55:28.874.491.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      29.110379 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.6123186 to 30.6122108
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.9998922 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220629_165456_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.3758535 to 2.3761425
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 29.1102710  -- OK for SL --
   MIU time from 2022-06-29T16:54:59.763.534.000 to 2022-06-29T16:54:59.763.823.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 29.110379

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4864135 to 31.4867025
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-29T16:55:28.874.202.000 to 2022-06-29T16:55:28.874.491.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.375853500 corrected by 40bit 10MHz counter of 1.612318600 obtained at 1Hz counter of 0. Result: 0.763537725
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996200 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-29T16:54:59.763.393.725
   Max 1st sample MIU time among all SL packets: 2022-06-29T16:55:28.874.350.725
   Min 1st sample XO time among all SL packets: 0.763397450
   Max 1st sample XO time among all SL packets: 29.874354748
   TM values from -218 to 202 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-29T16:54:59.763.393.725 (XO: 0.763397450 - 0.763703451, t=37.2 degC)  Rank: 00081
   TM values: Max =   202  Min =  -218  Ave =    -4.52  Std =    24.36
   Min TM value:-218 at an offset of 122.088 usec -> 2022-06-29T16:54:59.763.515.812 (XO: 0.763519538)
   Max TM value:202 at an offset of 124.213 usec -> 2022-06-29T16:54:59.763.517.937 (XO: 0.763521663)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 122.188 usec -> 2022-06-29T16:54:59.763.515.912 (XO: 0.763519638)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 124.325 usec -> 2022-06-29T16:54:59.763.518.050 (XO: 0.763521775)

 - Block number: 2 starting at 2022-06-29T16:55:28.874.061.725 (XO: 29.874065746 - 29.874371748, t=37.2 degC)  Rank: 00073
   TM values: Max =   198  Min =  -197  Ave =    -4.43  Std =    20.23
   Min TM value:-197 at an offset of 115.088 usec -> 2022-06-29T16:55:28.874.176.812 (XO: 29.874180834)
   Max TM value:198 at an offset of 112.600 usec -> 2022-06-29T16:55:28.874.174.325 (XO: 29.874178347)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 157.038 usec -> 2022-06-29T16:55:28.874.218.763 (XO: 29.874222785)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 112.700 usec -> 2022-06-29T16:55:28.874.174.425 (XO: 29.874178447)
