-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 02:06:30 2023 UTC
-- File 20220629_152549_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.597012 to 16.587603
-each separately from 8s to 16s and from 587314us to 597301us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.990013 in 1 packets: 18
MIU packet time from 2022-06-29T15:26:00.597.012.000 to 2022-06-29T15:26:08.587.603.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.990013 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.6440527 to 17.6440230
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999703 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220629_152549_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.2410655 to 10.2413545
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.9899830  -- OK for SL --
   MIU time from 2022-06-29T15:26:00.597.012.000 to 2022-06-29T15:26:00.597.301.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.990013

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.2313375 to 18.2316265
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-29T15:26:08.587.314.000 to 2022-06-29T15:26:08.587.603.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.241065500 corrected by 40bit 10MHz counter of 9.644052700 obtained at 1Hz counter of 8. Result: 8.597015009
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-29T15:26:00.596.871.725
   Max 1st sample MIU time among all SL packets: 2022-06-29T15:26:08.587.462.725
   Min 1st sample XO time among all SL packets: 8.596874733
   Max 1st sample XO time among all SL packets: 16.587465399
   TM values from -237 to 227 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-29T15:26:00.596.871.725 (XO: 8.596874733 - 8.597180735, t=35.6 degC)  Rank: 00082
   TM values: Max =   227  Min =  -215  Ave =    -4.49  Std =    25.23
   Min TM value:-215 at an offset of 118.413 usec -> 2022-06-29T15:26:00.596.990.137 (XO: 8.596993146)
   Max TM value:227 at an offset of 120.525 usec -> 2022-06-29T15:26:00.596.992.250 (XO: 8.596995259)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 120.438 usec -> 2022-06-29T15:26:00.596.992.162 (XO: 8.596995171)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 120.625 usec -> 2022-06-29T15:26:00.596.992.350 (XO: 8.596995359)

 - Block number: 2 starting at 2022-06-29T15:26:08.587.173.725 (XO: 16.587176398 - 16.587482399, t=35.6 degC)  Rank: 00079
   TM values: Max =   200  Min =  -237  Ave =    -4.58  Std =    22.93
   Min TM value:-237 at an offset of 95.725 usec -> 2022-06-29T15:26:08.587.269.450 (XO: 16.587272123)
   Max TM value:200 at an offset of 93.275 usec -> 2022-06-29T15:26:08.587.267.000 (XO: 16.587269673)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 95.850 usec -> 2022-06-29T15:26:08.587.269.575 (XO: 16.587272248)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 95.638 usec -> 2022-06-29T15:26:08.587.269.362 (XO: 16.587272035)
