-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 01:33:49 2023 UTC
-- File 20220629_145606_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.115290 to 81.138296
-each separately from 4s to 81s and from 115290us to 907970us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.330593 in 1 packets: 18
      7.939997 in 1 packets: 36
      22.980804 in 1 packets: 54
      5.539841 in 1 packets: 72
      18.230037 in 1 packets: 90
MIU packet time from 2022-06-29T14:56:13.115.290.000 to 2022-06-29T14:57:30.138.296.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.330593 in 1 packets: 18
      7.939997 in 1 packets: 36
      22.980804 in 1 packets: 54
      5.539841 in 1 packets: 72
      18.230037 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.7635951 to 82.7633105
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999187 in 1 packets: 18
      7.9999704 in 1 packets: 36
      22.9999151 in 1 packets: 54
      4.9999816 in 1 packets: 72
      18.9999296 in 1 packets: 90
Temperature FPGA: 
      05 -> 48.5 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Number of generated alerts: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20220629_145606_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.8788866 to 5.8791756
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.3305110  -- OK for SL --
   MIU time from 2022-06-29T14:56:13.115.290.000 to 2022-06-29T14:56:13.115.579.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.330593

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.2096866 to 28.2099756
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.9399670  -- OK for SL --
   MIU time from 2022-06-29T14:56:35.446.172.000 to 2022-06-29T14:56:35.446.461.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.939997

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.1499426 to 36.1502316
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.9807190  -- OK for SL --
   MIU time from 2022-06-29T14:56:43.386.458.000 to 2022-06-29T14:56:43.386.747.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.980804

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 59.1309506 to 59.1312396
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.5398230  -- OK for SL --
   MIU time from 2022-06-29T14:57:06.367.551.000 to 2022-06-29T14:57:06.367.840.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.539841

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 64.6710626 to 64.6713516
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.2299670  -- OK for SL --
   MIU time from 2022-06-29T14:57:11.907.681.000 to 2022-06-29T14:57:11.907.970.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.230037

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 82.9013186 to 82.9016076
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-29T14:57:30.138.007.000 to 2022-06-29T14:57:30.138.296.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 5.878886600 corrected by 40bit 10MHz counter of 5.763595100 obtained at 1Hz counter of 4. Result: 4.115291927
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996200 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-29T14:56:13.115.149.725
   Max 1st sample MIU time among all SL packets: 2022-06-29T14:57:30.138.155.725
   Min 1st sample XO time among all SL packets: 4.115151651
   Max 1st sample XO time among all SL packets: 81.138157336
   TM values from -243 to 226 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-29T14:56:13.115.149.725 (XO: 4.115151651 - 4.115457652, t=35.6 degC)  Rank: 00080
   TM values: Max =   194  Min =  -206  Ave =    -4.49  Std =    26.75
   Min TM value:-206 at an offset of 117.425 usec -> 2022-06-29T14:56:13.115.267.150 (XO: 4.115269076)
   Max TM value:194 at an offset of 122.475 usec -> 2022-06-29T14:56:13.115.272.200 (XO: 4.115274127)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 134.663 usec -> 2022-06-29T14:56:13.115.284.387 (XO: 4.115286314)
   Min dTM/dt [1/us]:-1.60e+03 at an offset of 134.838 usec -> 2022-06-29T14:56:13.115.284.562 (XO: 4.115286489)

 - Block number: 2 starting at 2022-06-29T14:56:35.446.031.725 (XO: 26.446034220 - 26.446340221, t=35.6 degC)  Rank: 00077
   TM values: Max =   210  Min =  -243  Ave =    -4.53  Std =    24.72
   Min TM value:-243 at an offset of 119.388 usec -> 2022-06-29T14:56:35.446.151.112 (XO: 26.446153608)
   Max TM value:210 at an offset of 116.925 usec -> 2022-06-29T14:56:35.446.148.650 (XO: 26.446151145)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 119.500 usec -> 2022-06-29T14:56:35.446.151.225 (XO: 26.446153720)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 119.300 usec -> 2022-06-29T14:56:35.446.151.025 (XO: 26.446153520)

 - Block number: 3 starting at 2022-06-29T14:56:43.386.317.725 (XO: 34.386319554 - 34.386625556, t=35.6 degC)  Rank: 00077
   TM values: Max =   220  Min =  -218  Ave =    -4.51  Std =    22.88
   Min TM value:-218 at an offset of 133.063 usec -> 2022-06-29T14:56:43.386.450.787 (XO: 34.386452617)
   Max TM value:220 at an offset of 130.600 usec -> 2022-06-29T14:56:43.386.448.325 (XO: 34.386450155)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 133.188 usec -> 2022-06-29T14:56:43.386.450.912 (XO: 34.386452742)
   Min dTM/dt [1/us]:-2.21e+03 at an offset of 130.688 usec -> 2022-06-29T14:56:43.386.448.412 (XO: 34.386450242)

 - Block number: 4 starting at 2022-06-29T14:57:06.367.410.725 (XO: 57.367412384 - 57.367718386, t=35.6 degC)  Rank: 00079
   TM values: Max =   169  Min =  -236  Ave =    -4.44  Std =    25.49
   Min TM value:-236 at an offset of 128.125 usec -> 2022-06-29T14:57:06.367.538.850 (XO: 57.367540510)
   Max TM value:169 at an offset of 133.175 usec -> 2022-06-29T14:57:06.367.543.900 (XO: 57.367545560)
   Max dTM/dt [1/us]:1.42e+03 at an offset of 161.063 usec -> 2022-06-29T14:57:06.367.571.788 (XO: 57.367573448)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 145.701 usec -> 2022-06-29T14:57:06.367.556.425 (XO: 57.367558085)

 - Block number: 5 starting at 2022-06-29T14:57:11.907.540.725 (XO: 62.907544783 - 62.907850784, t=35.6 degC)  Rank: 00087
   TM values: Max =   226  Min =  -241  Ave =    -4.44  Std =    19.39
   Min TM value:-241 at an offset of 120.613 usec -> 2022-06-29T14:57:11.907.661.337 (XO: 62.907665396)
   Max TM value:226 at an offset of 118.163 usec -> 2022-06-29T14:57:11.907.658.887 (XO: 62.907662946)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 120.713 usec -> 2022-06-29T14:57:11.907.661.437 (XO: 62.907665496)
   Min dTM/dt [1/us]:-2.32e+03 at an offset of 118.263 usec -> 2022-06-29T14:57:11.907.658.987 (XO: 62.907663046)

 - Block number: 6 starting at 2022-06-29T14:57:30.137.866.725 (XO: 81.137868335 - 81.138174336, t=35.6 degC)  Rank: 00088
   TM values: Max =   194  Min =  -220  Ave =    -4.35  Std =    18.95
   Min TM value:-220 at an offset of 119.350 usec -> 2022-06-29T14:57:30.137.986.075 (XO: 81.137987686)
   Max TM value:194 at an offset of 124.350 usec -> 2022-06-29T14:57:30.137.991.075 (XO: 81.137992686)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 117.188 usec -> 2022-06-29T14:57:30.137.983.912 (XO: 81.137985523)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 121.425 usec -> 2022-06-29T14:57:30.137.988.150 (XO: 81.137989761)
