-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 00:21:00 2023 UTC
-- File 20220629_135512_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.157046 to 44.477003
-each separately from 4s to 44s and from 157046us to 917315us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.759691 in 1 packets: 18
      23.559399 in 1 packets: 36
MIU packet time from 2022-06-29T13:55:19.157.046.000 to 2022-06-29T13:55:59.477.003.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.759691 in 1 packets: 18
      23.559399 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.2582373 to 46.2580899
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999411 in 1 packets: 18
      23.9999115 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220629_135512_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.4152844 to 6.4155734
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.7596310  -- OK for SL --
   MIU time from 2022-06-29T13:55:19.157.046.000 to 2022-06-29T13:55:19.157.335.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.759691

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.1752044 to 23.1754934
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.5593110  -- OK for SL --
   MIU time from 2022-06-29T13:55:35.917.026.000 to 2022-06-29T13:55:35.917.315.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.559399

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.7348044 to 46.7350934
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-29T13:55:59.476.714.000 to 2022-06-29T13:55:59.477.003.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.415284400 corrected by 40bit 10MHz counter of 6.258237300 obtained at 1Hz counter of 4. Result: 4.157047681
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-29T13:55:19.156.905.725
   Max 1st sample MIU time among all SL packets: 2022-06-29T13:55:59.476.862.725
   Min 1st sample XO time among all SL packets: 4.156907406
   Max 1st sample XO time among all SL packets: 44.476864989
   TM values from -236 to 186 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-29T13:55:19.156.905.725 (XO: 4.156907406 - 4.157213407, t=35.6 degC)  Rank: 00076
   TM values: Max =   163  Min =  -197  Ave =    -4.48  Std =    16.67
   Min TM value:-197 at an offset of 118.675 usec -> 2022-06-29T13:55:19.157.024.400 (XO: 4.157026081)
   Max TM value:163 at an offset of 118.313 usec -> 2022-06-29T13:55:19.157.024.037 (XO: 4.157025718)
   Max dTM/dt [1/us]:1.26e+03 at an offset of 121.000 usec -> 2022-06-29T13:55:19.157.026.725 (XO: 4.157028406)
   Min dTM/dt [1/us]:-1.76e+03 at an offset of 118.438 usec -> 2022-06-29T13:55:19.157.024.162 (XO: 4.157025843)

 - Block number: 2 starting at 2022-06-29T13:55:35.916.885.725 (XO: 20.916889117 - 20.917195119, t=35.6 degC)  Rank: 00085
   TM values: Max =   186  Min =  -229  Ave =    -4.50  Std =    20.02
   Min TM value:-229 at an offset of 118.400 usec -> 2022-06-29T13:55:35.917.004.125 (XO: 20.917007518)
   Max TM value:186 at an offset of 118.050 usec -> 2022-06-29T13:55:35.917.003.775 (XO: 20.917007168)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 120.688 usec -> 2022-06-29T13:55:35.917.006.412 (XO: 20.917009805)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 120.500 usec -> 2022-06-29T13:55:35.917.006.225 (XO: 20.917009618)

 - Block number: 3 starting at 2022-06-29T13:55:59.476.573.725 (XO: 44.476575988 - 44.476881989, t=35.6 degC)  Rank: 00082
   TM values: Max =   178  Min =  -236  Ave =    -4.49  Std =    21.97
   Min TM value:-236 at an offset of 119.738 usec -> 2022-06-29T13:55:59.476.693.462 (XO: 44.476695726)
   Max TM value:178 at an offset of 119.425 usec -> 2022-06-29T13:55:59.476.693.150 (XO: 44.476695414)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 122.013 usec -> 2022-06-29T13:55:59.476.695.737 (XO: 44.476698001)
   Min dTM/dt [1/us]:-2.03e+03 at an offset of 121.825 usec -> 2022-06-29T13:55:59.476.695.550 (XO: 44.476697814)
