-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 00:07:32 2023 UTC
-- File 20220629_134619_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.169735 to 23.110260
-each separately from 11s to 23s and from 109971us to 170024us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.939947 in 1 packets: 18
MIU packet time from 2022-06-29T13:46:33.169.735.000 to 2022-06-29T13:46:45.110.260.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.939947 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.6163603 to 24.6163161
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999558 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      C0 -> 35.6 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220629_134619_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.7860965 to 12.7863855
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.9399030  -- OK for SL --
   MIU time from 2022-06-29T13:46:33.169.735.000 to 2022-06-29T13:46:33.170.024.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.939947

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.7262885 to 24.7265775
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-29T13:46:45.109.971.000 to 2022-06-29T13:46:45.110.260.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 12.786096500 corrected by 40bit 10MHz counter of 12.616360300 obtained at 1Hz counter of 11. Result: 11.169736828
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-29T13:46:33.169.594.725
   Max 1st sample MIU time among all SL packets: 2022-06-29T13:46:45.110.119.725
   Min 1st sample XO time among all SL packets: 11.169596553
   Max 1st sample XO time among all SL packets: 23.110121532
   TM values from -217 to 201 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-29T13:46:33.169.594.725 (XO: 11.169596553 - 11.169902554, t=37.2 degC)  Rank: 00080
   TM values: Max =   198  Min =  -211  Ave =    -4.54  Std =    20.63
   Min TM value:-211 at an offset of 110.913 usec -> 2022-06-29T13:46:33.169.705.637 (XO: 11.169707465)
   Max TM value:198 at an offset of 108.438 usec -> 2022-06-29T13:46:33.169.703.162 (XO: 11.169704990)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 111.025 usec -> 2022-06-29T13:46:33.169.705.750 (XO: 11.169707578)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 108.550 usec -> 2022-06-29T13:46:33.169.703.275 (XO: 11.169705103)

 - Block number: 2 starting at 2022-06-29T13:46:45.109.830.725 (XO: 23.109832531 - 23.110138533, t=35.6 degC)  Rank: 00077
   TM values: Max =   201  Min =  -217  Ave =    -4.50  Std =    19.81
   Min TM value:-217 at an offset of 99.425 usec -> 2022-06-29T13:46:45.109.930.150 (XO: 23.109931957)
   Max TM value:201 at an offset of 96.963 usec -> 2022-06-29T13:46:45.109.927.687 (XO: 23.109929494)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 99.588 usec -> 2022-06-29T13:46:45.109.930.312 (XO: 23.109932119)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 97.075 usec -> 2022-06-29T13:46:45.109.927.800 (XO: 23.109929607)
