-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 23:59:55 2023 UTC
-- File 20220629_133859_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.725785 to 22.766187
-each separately from 13s to 22s and from 725785us to 766187us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.039824 in 1 packets: 18
MIU packet time from 2022-06-29T13:39:15.725.785.000 to 2022-06-29T13:39:24.766.187.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.039824 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.2561801 to 24.2561470
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999669 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220629_133859_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.9819652 to 15.9822542
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.0397910  -- OK for SL --
   MIU time from 2022-06-29T13:39:15.725.785.000 to 2022-06-29T13:39:15.726.074.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.039824

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.0220452 to 25.0223342
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-29T13:39:24.765.898.000 to 2022-06-29T13:39:24.766.187.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.981965200 corrected by 40bit 10MHz counter of 15.256180100 obtained at 1Hz counter of 13. Result: 13.725787785
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-29T13:39:15.725.644.725
   Max 1st sample MIU time among all SL packets: 2022-06-29T13:39:24.766.046.725
   Min 1st sample XO time among all SL packets: 13.725647510
   Max 1st sample XO time among all SL packets: 22.766049759
   TM values from -233 to 208 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-29T13:39:15.725.644.725 (XO: 13.725647510 - 13.725953511, t=35.6 degC)  Rank: 00090
   TM values: Max =   208  Min =  -233  Ave =    -4.43  Std =    21.28
   Min TM value:-233 at an offset of 115.825 usec -> 2022-06-29T13:39:15.725.760.550 (XO: 13.725763335)
   Max TM value:208 at an offset of 113.375 usec -> 2022-06-29T13:39:15.725.758.100 (XO: 13.725760885)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 115.938 usec -> 2022-06-29T13:39:15.725.760.662 (XO: 13.725763448)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 115.738 usec -> 2022-06-29T13:39:15.725.760.462 (XO: 13.725763248)

 - Block number: 2 starting at 2022-06-29T13:39:24.765.757.725 (XO: 22.765760758 - 22.766066759, t=35.6 degC)  Rank: 00080
   TM values: Max =   168  Min =  -224  Ave =    -4.54  Std =    19.50
   Min TM value:-224 at an offset of 126.438 usec -> 2022-06-29T13:39:24.765.884.162 (XO: 22.765887196)
   Max TM value:168 at an offset of 126.100 usec -> 2022-06-29T13:39:24.765.883.825 (XO: 22.765886859)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 128.725 usec -> 2022-06-29T13:39:24.765.886.450 (XO: 22.765889484)
   Min dTM/dt [1/us]:-1.72e+03 at an offset of 114.425 usec -> 2022-06-29T13:39:24.765.872.150 (XO: 22.765875184)
