-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 23:39:25 2023 UTC
-- File 20220629_132111_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.172402 to 65.692514
-each separately from 9s to 65s and from 62311us to 802320us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.889620 in 1 packets: 18
      24.739431 in 1 packets: 36
      3.649742 in 1 packets: 54
      18.239874 in 1 packets: 72
MIU packet time from 2022-06-29T13:21:23.172.402.000 to 2022-06-29T13:22:19.692.514.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.889620 in 1 packets: 18
      24.739431 in 1 packets: 36
      3.649742 in 1 packets: 54
      18.239874 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.1188130 to 67.1186072
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999632 in 1 packets: 18
      23.9999117 in 1 packets: 36
      3.9999854 in 1 packets: 54
      17.9999339 in 1 packets: 72
Temperature FPGA: 
      06 -> 47.2 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C0 -> 35.6 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220629_132111_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.2912158 to 11.2915048
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.8895830  -- OK for SL --
   MIU time from 2022-06-29T13:21:23.172.402.000 to 2022-06-29T13:21:23.172.691.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.889620

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.1810878 to 21.1813768
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.7393430  -- OK for SL --
   MIU time from 2022-06-29T13:21:33.062.311.000 to 2022-06-29T13:21:33.062.600.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.739431

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.9207198 to 45.9210088
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.6497270  -- OK for SL --
   MIU time from 2022-06-29T13:21:57.802.031.000 to 2022-06-29T13:21:57.802.320.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.649742

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.5707358 to 49.5710248
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.2398070  -- OK for SL --
   MIU time from 2022-06-29T13:22:01.452.062.000 to 2022-06-29T13:22:01.452.351.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.239874

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 67.8108318 to 67.8111208
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-29T13:22:19.692.225.000 to 2022-06-29T13:22:19.692.514.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.291215800 corrected by 40bit 10MHz counter of 11.118813000 obtained at 1Hz counter of 9. Result: 9.172403438
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-06-29T13:21:23.172.261.725
   Max 1st sample MIU time among all SL packets: 2022-06-29T13:22:19.692.373.725
   Min 1st sample XO time among all SL packets: 9.172263162
   Max 1st sample XO time among all SL packets: 65.692375887
   TM values from -249 to 219 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-29T13:21:23.172.261.725 (XO: 9.172263162 - 9.172569164, t=35.6 degC)  Rank: 00077
   TM values: Max =   206  Min =  -244  Ave =    -4.55  Std =    20.16
   Min TM value:-244 at an offset of 102.088 usec -> 2022-06-29T13:21:23.172.363.812 (XO: 9.172365250)
   Max TM value:206 at an offset of 99.625 usec -> 2022-06-29T13:21:23.172.361.350 (XO: 9.172362788)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 102.188 usec -> 2022-06-29T13:21:23.172.363.912 (XO: 9.172365350)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 101.988 usec -> 2022-06-29T13:21:23.172.363.712 (XO: 9.172365150)

 - Block number: 2 starting at 2022-06-29T13:21:33.062.170.725 (XO: 19.062171549 - 19.062477550, t=35.6 degC)  Rank: 00080
   TM values: Max =   219  Min =  -213  Ave =    -4.44  Std =    18.15
   Min TM value:-213 at an offset of 116.063 usec -> 2022-06-29T13:21:33.062.286.787 (XO: 19.062287612)
   Max TM value:219 at an offset of 113.600 usec -> 2022-06-29T13:21:33.062.284.325 (XO: 19.062285149)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 113.525 usec -> 2022-06-29T13:21:33.062.284.250 (XO: 19.062285074)
   Min dTM/dt [1/us]:-1.93e+03 at an offset of 113.700 usec -> 2022-06-29T13:21:33.062.284.425 (XO: 19.062285249)

 - Block number: 3 starting at 2022-06-29T13:21:57.801.890.725 (XO: 43.801894512 - 43.802200513, t=35.6 degC)  Rank: 00072
   TM values: Max =   201  Min =  -215  Ave =    -4.30  Std =    18.54
   Min TM value:-215 at an offset of 114.575 usec -> 2022-06-29T13:21:57.802.005.300 (XO: 43.802009087)
   Max TM value:201 at an offset of 112.100 usec -> 2022-06-29T13:21:57.802.002.825 (XO: 43.802006612)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 114.700 usec -> 2022-06-29T13:21:57.802.005.425 (XO: 43.802009212)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 112.238 usec -> 2022-06-29T13:21:57.802.002.962 (XO: 43.802006750)

 - Block number: 4 starting at 2022-06-29T13:22:01.451.921.725 (XO: 47.451923897 - 47.452229898, t=35.6 degC)  Rank: 00078
   TM values: Max =   180  Min =  -249  Ave =    -4.47  Std =    22.51
   Min TM value:-249 at an offset of 110.963 usec -> 2022-06-29T13:22:01.452.032.687 (XO: 47.452034860)
   Max TM value:180 at an offset of 68.275 usec -> 2022-06-29T13:22:01.451.990.000 (XO: 47.451992172)
   Max dTM/dt [1/us]:1.47e+03 at an offset of 68.200 usec -> 2022-06-29T13:22:01.451.989.925 (XO: 47.451992097)
   Min dTM/dt [1/us]:-1.49e+03 at an offset of 68.363 usec -> 2022-06-29T13:22:01.451.990.087 (XO: 47.451992260)

 - Block number: 5 starting at 2022-06-29T13:22:19.692.084.725 (XO: 65.692086886 - 65.692392887, t=35.6 degC)  Rank: 00083
   TM values: Max =   208  Min =  -239  Ave =    -4.58  Std =    21.63
   Min TM value:-239 at an offset of 91.163 usec -> 2022-06-29T13:22:19.692.175.887 (XO: 65.692178049)
   Max TM value:208 at an offset of 88.700 usec -> 2022-06-29T13:22:19.692.173.425 (XO: 65.692175586)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 91.275 usec -> 2022-06-29T13:22:19.692.176.000 (XO: 65.692178161)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 91.075 usec -> 2022-06-29T13:22:19.692.175.800 (XO: 65.692177961)
