-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar 21 00:27:01 2023 UTC
-- File 20220628_080910_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.315158 to 27.666335
-each separately from 12s to 27s and from 315158us to 666335us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.350599 in 1 packets: 18
MIU packet time from 2022-06-28T08:09:25.315.158.000 to 2022-06-28T08:09:40.666.335.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.350599 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.1509650 to 29.1509096
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999446 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220628_080910_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.4661238 to 14.4664128
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.3505430  -- OK for SL --
   MIU time from 2022-06-28T08:09:25.315.158.000 to 2022-06-28T08:09:25.315.447.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.350599

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      113 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.8169558 to 29.8172448
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-28T08:09:40.666.046.000 to 2022-06-28T08:09:40.666.335.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.466123800 corrected by 40bit 10MHz counter of 14.150965000 obtained at 1Hz counter of 12. Result: 12.315159966
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-28T08:09:25.315.017.725
   Max 1st sample MIU time among all SL packets: 2022-06-28T08:09:40.666.194.725
   Min 1st sample XO time among all SL packets: 12.315019691
   Max 1st sample XO time among all SL packets: 27.666197390
   TM values from -680 to 325 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-28T08:09:25.315.017.725 (XO: 12.315019691 - 12.315325692, t=37.2 degC)  Rank: 00078
   TM values: Max =   263  Min =  -680  Ave =    -4.51  Std =    31.63
   Min TM value:-680 at an offset of 57.788 usec -> 2022-06-28T08:09:25.315.075.512 (XO: 12.315077478)
   Max TM value:263 at an offset of 47.638 usec -> 2022-06-28T08:09:25.315.065.362 (XO: 12.315067328)
   Max dTM/dt [1/us]:6.56e+03 at an offset of 47.588 usec -> 2022-06-28T08:09:25.315.065.312 (XO: 12.315067278)
   Min dTM/dt [1/us]:-5.59e+03 at an offset of 57.738 usec -> 2022-06-28T08:09:25.315.075.462 (XO: 12.315077428)

 - Block number: 2 starting at 2022-06-28T08:09:40.665.905.725 (XO: 27.665908389 - 27.666214390, t=37.2 degC)  Rank: 00113
   TM values: Max =   325  Min =  -388  Ave =    -4.56  Std =    53.83
   Min TM value:-388 at an offset of 95.100 usec -> 2022-06-28T08:09:40.666.000.825 (XO: 27.666003489)
   Max TM value:325 at an offset of 102.300 usec -> 2022-06-28T08:09:40.666.008.025 (XO: 27.666010689)
   Max dTM/dt [1/us]:5.03e+03 at an offset of 95.175 usec -> 2022-06-28T08:09:40.666.000.900 (XO: 27.666003564)
   Min dTM/dt [1/us]:-2.96e+03 at an offset of 246.451 usec -> 2022-06-28T08:09:40.666.152.175 (XO: 27.666154840)
