-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar 21 00:18:46 2023 UTC
-- File 20220628_080355_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.285326 to 30.008406
-each separately from 2s to 30s and from 8117us to 285615us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.722502 in 1 packets: 18
MIU packet time from 2022-06-28T08:04:00.285.326.000 to 2022-06-28T08:04:28.008.406.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.722502 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.7671215 to 31.7670181
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.9998966 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 1 packets: 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 1 packets: 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220628_080355_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.0524483 to 4.0527373
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 27.7223990  -- OK for SL --
   MIU time from 2022-06-28T08:04:00.285.326.000 to 2022-06-28T08:04:00.285.615.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 27.722502

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.7751363 to 31.7754253
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-28T08:04:28.008.117.000 to 2022-06-28T08:04:28.008.406.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.052448300 corrected by 40bit 10MHz counter of 3.767121500 obtained at 1Hz counter of 2. Result: 2.285327856
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-28T08:04:00.285.185.725
   Max 1st sample MIU time among all SL packets: 2022-06-28T08:04:28.008.265.725
   Min 1st sample XO time among all SL packets: 2.285187580
   Max 1st sample XO time among all SL packets: 30.008267056
   TM values from -241 to 195 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-28T08:04:00.285.185.725 (XO: 2.285187580 - 2.285493581, t=37.2 degC)  Rank: 00081
   TM values: Max =   175  Min =  -238  Ave =    -4.39  Std =    18.78
   Min TM value:-238 at an offset of 121.238 usec -> 2022-06-28T08:04:00.285.306.962 (XO: 2.285308818)
   Max TM value:175 at an offset of 118.713 usec -> 2022-06-28T08:04:00.285.304.437 (XO: 2.285306293)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 121.338 usec -> 2022-06-28T08:04:00.285.307.062 (XO: 2.285308918)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 118.825 usec -> 2022-06-28T08:04:00.285.304.550 (XO: 2.285306406)

 - Block number: 2 starting at 2022-06-28T08:04:28.007.976.725 (XO: 30.007978055 - 30.008284056, t=37.2 degC)  Rank: 00086
   TM values: Max =   195  Min =  -241  Ave =    -4.49  Std =    27.65
   Min TM value:-241 at an offset of 123.488 usec -> 2022-06-28T08:04:28.008.100.212 (XO: 30.008101542)
   Max TM value:195 at an offset of 120.988 usec -> 2022-06-28T08:04:28.008.097.712 (XO: 30.008099042)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 123.613 usec -> 2022-06-28T08:04:28.008.100.337 (XO: 30.008101667)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 121.088 usec -> 2022-06-28T08:04:28.008.097.812 (XO: 30.008099142)
