-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar 21 00:10:14 2023 UTC
-- File 20220628_074311_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.465179 to 19.044215
-each separately from 3s to 19s and from 43926us to 465468us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.578458 in 1 packets: 18
MIU packet time from 2022-06-28T07:43:17.465.179.000 to 2022-06-28T07:43:33.044.215.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.578458 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.7604723 to 20.7604126
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999403 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220628_074311_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.2256516 to 5.2259406
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.5783990  -- OK for SL --
   MIU time from 2022-06-28T07:43:17.465.179.000 to 2022-06-28T07:43:17.465.468.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.578458

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.8043396 to 20.8046286
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-28T07:43:33.043.926.000 to 2022-06-28T07:43:33.044.215.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.225651600 corrected by 40bit 10MHz counter of 4.760472300 obtained at 1Hz counter of 3. Result: 3.465181068
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996200 - 0.999996200
   Min 1st sample MIU time among all SL packets: 2022-06-28T07:43:17.465.038.725
   Max 1st sample MIU time among all SL packets: 2022-06-28T07:43:33.044.074.725
   Min 1st sample XO time among all SL packets: 3.465040792
   Max 1st sample XO time among all SL packets: 19.044075892
   TM values from -290 to 254 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-28T07:43:17.465.038.725 (XO: 3.465040792 - 3.465346793, t=37.2 degC)  Rank: 00085
   TM values: Max =   254  Min =  -290  Ave =    -4.53  Std =    21.56
   Min TM value:-290 at an offset of 124.913 usec -> 2022-06-28T07:43:17.465.163.637 (XO: 3.465165705)
   Max TM value:254 at an offset of 122.450 usec -> 2022-06-28T07:43:17.465.161.175 (XO: 3.465163243)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 122.400 usec -> 2022-06-28T07:43:17.465.161.125 (XO: 3.465163193)
   Min dTM/dt [1/us]:-2.85e+03 at an offset of 122.563 usec -> 2022-06-28T07:43:17.465.161.287 (XO: 3.465163355)

 - Block number: 2 starting at 2022-06-28T07:43:33.043.785.725 (XO: 19.043786891 - 19.044092893, t=37.2 degC)  Rank: 00072
   TM values: Max =   219  Min =  -223  Ave =    -4.47  Std =    17.35
   Min TM value:-223 at an offset of 113.150 usec -> 2022-06-28T07:43:33.043.898.875 (XO: 19.043900042)
   Max TM value:219 at an offset of 112.863 usec -> 2022-06-28T07:43:33.043.898.587 (XO: 19.043899754)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 115.500 usec -> 2022-06-28T07:43:33.043.901.225 (XO: 19.043902392)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 112.975 usec -> 2022-06-28T07:43:33.043.898.700 (XO: 19.043899867)
