-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Mar 20 19:28:54 2023 UTC
-- File 20220628_034110_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177031
-each separately from 0s to 14s and from 0us to 841257us.
   Differences in neigboring packets: 
      0.833940 in 2 packets: 1, 7
      0.833943 in 6 packets: 2, 4, 10, 12, 15, 17
      0.833944 in 8 packets: 3, 5, 6, 8, 9, 11, 14, 16
      0.833941 in 1 packets: 13
MIU packet time from 2022-06-28T03:41:13.000.000.000 to 2022-06-28T03:41:27.177.031.000
   Differences in neigboring packets: 
      0.833940 in 2 packets: 1, 7
      0.833943 in 6 packets: 2, 4, 10, 12, 15, 17
      0.833944 in 8 packets: 3, 5, 6, 8, 9, 11, 14, 16
      0.833941 in 1 packets: 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.1153435 to 16.1152913
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999962 in 4 packets: 2, 6, 10, 15
      0.9999963 in 10 packets: 3, 4, 5, 8, 9, 11, 12, 14, 16, 17
   XO PPM: 
      -3.8 in 4 packets: 2, 6, 10, 15
      -3.7 in 10 packets: 3, 4, 5, 8, 9, 11, 12, 14, 16, 17
Temperature FPGA: 
      05 -> 48.5 degC in 12 packets: 1, 3, 4, 5, 9, 11, 12, 13, 14, 15, 16, 18
      04 -> 49.8 degC in 6 packets: 2, 6, 7, 8, 10, 17
Temperature XO: 
      C1 -> 37.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20220628_034110_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.1153436 to 16.2923236
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-28T03:41:13.000.000.000 to 2022-06-28T03:41:27.177.031.000
      Difference between neigboring packets: 
      0.833940 in 2 packets: 1, 7
      0.833943 in 6 packets: 2, 4, 10, 12, 15, 17
      0.833944 in 8 packets: 3, 5, 6, 8, 9, 11, 14, 16
      0.833941 in 1 packets: 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.115343600 corrected by 40bit 10MHz counter of 2.115343500 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996200 - 0.999996300
   Min 1st sample MIU time among all SR packets: 2022-06-28T03:41:12.999.999.925
   Max 1st sample MIU time among all SR packets: 2022-06-28T03:41:27.177.030.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177032880
   TM values from -184 to 170 median:-5.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2022-06-28T03:41:12.999.999.925 (XO: 0.000000025 - 0.000017025, t=37.2 degC)
   TM values: Max =    14  Min =   -22  Ave =    -4.24  Std =     6.44

 - Packet number: 2 starting at 2022-06-28T03:41:13.833.939.925 (XO: 0.833943194 - 0.833960194, t=37.2 degC)
   TM values: Max =    15  Min =   -25  Ave =    -4.50  Std =     7.94

 - Packet number: 3 starting at 2022-06-28T03:41:14.667.882.925 (XO: 1.667886296 - 1.667903296, t=37.2 degC)
   TM values: Max =    12  Min =   -21  Ave =    -4.49  Std =     5.76

 - Packet number: 4 starting at 2022-06-28T03:41:15.501.826.925 (XO: 2.501829382 - 2.501846382, t=37.2 degC)
   TM values: Max =    16  Min =   -22  Ave =    -4.24  Std =     6.99

 - Packet number: 5 starting at 2022-06-28T03:41:16.335.769.925 (XO: 3.335772467 - 3.335789467, t=37.2 degC)
   TM values: Max =    10  Min =   -17  Ave =    -4.52  Std =     4.54

 - Packet number: 6 starting at 2022-06-28T03:41:17.169.713.925 (XO: 4.169715570 - 4.169732570, t=37.2 degC)
   TM values: Max =    13  Min =   -25  Ave =    -4.31  Std =     6.43

 - Packet number: 7 starting at 2022-06-28T03:41:18.003.657.925 (XO: 5.003658739 - 5.003675739, t=37.2 degC)
   TM values: Max =    12  Min =   -20  Ave =    -4.57  Std =     5.80

 - Packet number: 8 starting at 2022-06-28T03:41:18.837.597.925 (XO: 5.837601824 - 5.837618824, t=37.2 degC)
   TM values: Max =     9  Min =   -20  Ave =    -4.51  Std =     5.87

 - Packet number: 9 starting at 2022-06-28T03:41:19.671.541.925 (XO: 6.671544910 - 6.671561910, t=37.2 degC)
   TM values: Max =    10  Min =   -19  Ave =    -4.56  Std =     5.65

 - Packet number: 10 starting at 2022-06-28T03:41:20.505.485.925 (XO: 7.505488046 - 7.505505046, t=37.2 degC)
   TM values: Max =     7  Min =   -15  Ave =    -4.43  Std =     3.66

 - Packet number: 11 starting at 2022-06-28T03:41:21.339.428.925 (XO: 8.339431181 - 8.339448181, t=37.2 degC)
   TM values: Max =    12  Min =   -21  Ave =    -4.64  Std =     5.86

 - Packet number: 12 starting at 2022-06-28T03:41:22.173.372.925 (XO: 9.173374266 - 9.173391267, t=37.2 degC)
   TM values: Max =     9  Min =   -18  Ave =    -4.46  Std =     5.52

 - Packet number: 13 starting at 2022-06-28T03:41:23.007.315.925 (XO: 10.007317352 - 10.007334352, t=37.2 degC)
   TM values: Max =    12  Min =   -19  Ave =    -4.60  Std =     5.21

 - Packet number: 14 starting at 2022-06-28T03:41:23.841.256.925 (XO: 10.841260438 - 10.841277438, t=37.2 degC)
   TM values: Max =    13  Min =   -21  Ave =    -4.53  Std =     7.10

 - Packet number: 15 starting at 2022-06-28T03:41:24.675.200.925 (XO: 11.675203591 - 11.675220591, t=37.2 degC)
   TM values: Max =   170  Min =  -184  Ave =    -4.40  Std =    34.73

 - Packet number: 16 starting at 2022-06-28T03:41:25.509.143.925 (XO: 12.509146709 - 12.509163709, t=37.2 degC)
   TM values: Max =    10  Min =   -19  Ave =    -4.61  Std =     5.09

 - Packet number: 17 starting at 2022-06-28T03:41:26.343.087.925 (XO: 13.343089794 - 13.343106794, t=37.2 degC)
   TM values: Max =     7  Min =   -19  Ave =    -4.76  Std =     4.37

 - Packet number: 18 starting at 2022-06-28T03:41:27.177.030.925 (XO: 14.177032880 - 14.177049880, t=37.2 degC)
   TM values: Max =    11  Min =   -22  Ave =    -4.66  Std =     5.71
