-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Mar 20 18:31:48 2023 UTC
-- File 20220628_030715_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.621613 to 22.021850
-each separately from 10s to 22s and from 21561us to 621902us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.399659 in 1 packets: 18
MIU packet time from 2022-06-28T03:07:28.621.613.000 to 2022-06-28T03:07:40.021.850.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.399659 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.7674985 to 23.7674537
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999552 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220628_030715_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.3891117 to 12.3894007
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.3996150  -- OK for SL --
   MIU time from 2022-06-28T03:07:28.621.613.000 to 2022-06-28T03:07:28.621.902.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.399659

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.7890157 to 23.7893047
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-28T03:07:40.021.561.000 to 2022-06-28T03:07:40.021.850.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.389111700 corrected by 40bit 10MHz counter of 11.767498500 obtained at 1Hz counter of 10. Result: 10.621615500
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-28T03:07:28.621.472.725
   Max 1st sample MIU time among all SL packets: 2022-06-28T03:07:40.021.709.725
   Min 1st sample XO time among all SL packets: 10.621475224
   Max 1st sample XO time among all SL packets: 22.021710805
   TM values from -247 to 174 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-28T03:07:28.621.472.725 (XO: 10.621475224 - 10.621781226, t=37.2 degC)  Rank: 00087
   TM values: Max =   165  Min =  -209  Ave =    -4.56  Std =    25.69
   Min TM value:-209 at an offset of 109.438 usec -> 2022-06-28T03:07:28.621.582.162 (XO: 10.621584662)
   Max TM value:165 at an offset of 112.075 usec -> 2022-06-28T03:07:28.621.584.800 (XO: 10.621587300)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 139.463 usec -> 2022-06-28T03:07:28.621.612.188 (XO: 10.621614687)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 106.938 usec -> 2022-06-28T03:07:28.621.579.662 (XO: 10.621582162)

 - Block number: 2 starting at 2022-06-28T03:07:40.021.420.725 (XO: 22.021421804 - 22.021727805, t=37.2 degC)  Rank: 00081
   TM values: Max =   174  Min =  -247  Ave =    -4.48  Std =    25.76
   Min TM value:-247 at an offset of 162.776 usec -> 2022-06-28T03:07:40.021.583.500 (XO: 22.021584580)
   Max TM value:174 at an offset of 163.663 usec -> 2022-06-28T03:07:40.021.584.388 (XO: 22.021585467)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 165.163 usec -> 2022-06-28T03:07:40.021.585.888 (XO: 22.021586967)
   Min dTM/dt [1/us]:-1.72e+03 at an offset of 162.601 usec -> 2022-06-28T03:07:40.021.583.325 (XO: 22.021584405)
