-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 19:56:45 2023 UTC
-- File 20220624_164721_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.344113 to 17.864944
-each separately from 0s to 17s and from 344113us to 864944us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.520253 in 1 packets: 18
MIU packet time from 2022-06-24T16:47:24.344.113.000 to 2022-06-24T16:47:41.864.944.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.520253 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.1222827 to 19.1222207
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999380 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220624_164721_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.4663966 to 2.4666856
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.5201910  -- OK for SL --
   MIU time from 2022-06-24T16:47:24.344.113.000 to 2022-06-24T16:47:24.344.402.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.520253

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.9868766 to 19.9871656
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-24T16:47:41.864.655.000 to 2022-06-24T16:47:41.864.944.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.466396600 corrected by 40bit 10MHz counter of 2.122282700 obtained at 1Hz counter of 0. Result: 0.344115173
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-24T16:47:24.343.972.725
   Max 1st sample MIU time among all SL packets: 2022-06-24T16:47:41.864.803.725
   Min 1st sample XO time among all SL packets: 0.343974898
   Max 1st sample XO time among all SL packets: 17.864807825
   TM values from -307 to 229 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-24T16:47:24.343.972.725 (XO: 0.343974898 - 0.344280899, t=37.2 degC)  Rank: 00083
   TM values: Max =   213  Min =  -229  Ave =    -4.43  Std =    30.59
   Min TM value:-229 at an offset of 196.138 usec -> 2022-06-24T16:47:24.344.168.863 (XO: 0.344171036)
   Max TM value:213 at an offset of 195.851 usec -> 2022-06-24T16:47:24.344.168.575 (XO: 0.344170748)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 198.526 usec -> 2022-06-24T16:47:24.344.171.250 (XO: 0.344173423)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 95.375 usec -> 2022-06-24T16:47:24.344.068.100 (XO: 0.344070273)

 - Block number: 2 starting at 2022-06-24T16:47:41.864.514.725 (XO: 17.864518824 - 17.864824825, t=37.2 degC)  Rank: 00089
   TM values: Max =   229  Min =  -307  Ave =    -4.46  Std =    22.53
   Min TM value:-307 at an offset of 112.513 usec -> 2022-06-24T16:47:41.864.627.237 (XO: 17.864631337)
   Max TM value:229 at an offset of 108.288 usec -> 2022-06-24T16:47:41.864.623.012 (XO: 17.864627112)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 112.650 usec -> 2022-06-24T16:47:41.864.627.375 (XO: 17.864631474)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 110.213 usec -> 2022-06-24T16:47:41.864.624.937 (XO: 17.864629037)
