-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 16:26:54 2023 UTC
-- File 20220624_121635_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.156349 to 15.856271
-each separately from 2s to 15s and from 156349us to 856271us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.699344 in 1 packets: 18
MIU packet time from 2022-06-24T12:16:40.156.349.000 to 2022-06-24T12:16:53.856.271.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.699344 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.7591034 to 16.7590554
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999520 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220624_121635_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.9154533 to 3.9157423
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.6992950  -- OK for SL --
   MIU time from 2022-06-24T12:16:40.156.349.000 to 2022-06-24T12:16:40.156.638.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.699344

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.6150373 to 17.6153263
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-24T12:16:53.855.982.000 to 2022-06-24T12:16:53.856.271.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 3.915453300 corrected by 40bit 10MHz counter of 3.759103400 obtained at 1Hz counter of 2. Result: 2.156350478
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-24T12:16:40.156.208.725
   Max 1st sample MIU time among all SL packets: 2022-06-24T12:16:53.856.130.725
   Min 1st sample XO time among all SL packets: 2.156210203
   Max 1st sample XO time among all SL packets: 15.856133793
   TM values from -261 to 224 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-24T12:16:40.156.208.725 (XO: 2.156210203 - 2.156516204, t=35.6 degC)  Rank: 00076
   TM values: Max =   224  Min =  -261  Ave =    -4.45  Std =    21.27
   Min TM value:-261 at an offset of 124.300 usec -> 2022-06-24T12:16:40.156.333.025 (XO: 2.156334503)
   Max TM value:224 at an offset of 121.763 usec -> 2022-06-24T12:16:40.156.330.487 (XO: 2.156331966)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 124.400 usec -> 2022-06-24T12:16:40.156.333.125 (XO: 2.156334603)
   Min dTM/dt [1/us]:-2.38e+03 at an offset of 121.863 usec -> 2022-06-24T12:16:40.156.330.587 (XO: 2.156332066)

 - Block number: 2 starting at 2022-06-24T12:16:53.855.841.725 (XO: 15.855844792 - 15.856150793, t=35.6 degC)  Rank: 00083
   TM values: Max =   159  Min =  -206  Ave =    -4.51  Std =    26.03
   Min TM value:-206 at an offset of 123.588 usec -> 2022-06-24T12:16:53.855.965.312 (XO: 15.855968380)
   Max TM value:159 at an offset of 126.225 usec -> 2022-06-24T12:16:53.855.967.950 (XO: 15.855971017)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 123.688 usec -> 2022-06-24T12:16:53.855.965.412 (XO: 15.855968480)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 138.876 usec -> 2022-06-24T12:16:53.855.980.600 (XO: 15.855983667)
