-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 16:20:52 2023 UTC
-- File 20220624_121236_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.944380 to 23.403918
-each separately from 5s to 23s and from 403629us to 944669us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.458960 in 1 packets: 18
MIU packet time from 2022-06-24T12:12:44.944.380.000 to 2022-06-24T12:13:02.403.918.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.458960 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.7751147 to 24.7750483
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999336 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220624_121236_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.7194944 to 7.7197834
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.4588950  -- OK for SL --
   MIU time from 2022-06-24T12:12:44.944.380.000 to 2022-06-24T12:12:44.944.669.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.458960

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.1786784 to 25.1789674
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-24T12:13:02.403.629.000 to 2022-06-24T12:13:02.403.918.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.719494400 corrected by 40bit 10MHz counter of 6.775114700 obtained at 1Hz counter of 5. Result: 5.944383194
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-06-24T12:12:44.944.239.725
   Max 1st sample MIU time among all SL packets: 2022-06-24T12:13:02.403.777.725
   Min 1st sample XO time among all SL packets: 5.944242919
   Max 1st sample XO time among all SL packets: 23.403780279
   TM values from -274 to 204 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-24T12:12:44.944.239.725 (XO: 5.944242919 - 5.944548920, t=35.6 degC)  Rank: 00083
   TM values: Max =   168  Min =  -234  Ave =    -4.34  Std =    23.06
   Min TM value:-234 at an offset of 116.950 usec -> 2022-06-24T12:12:44.944.356.675 (XO: 5.944359869)
   Max TM value:168 at an offset of 121.825 usec -> 2022-06-24T12:12:44.944.361.550 (XO: 5.944364744)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 119.500 usec -> 2022-06-24T12:12:44.944.359.225 (XO: 5.944362419)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 122.013 usec -> 2022-06-24T12:12:44.944.361.737 (XO: 5.944364932)

 - Block number: 2 starting at 2022-06-24T12:13:02.403.488.725 (XO: 23.403491278 - 23.403797279, t=35.6 degC)  Rank: 00076
   TM values: Max =   204  Min =  -274  Ave =    -4.46  Std =    18.56
   Min TM value:-274 at an offset of 118.325 usec -> 2022-06-24T12:13:02.403.607.050 (XO: 23.403609603)
   Max TM value:204 at an offset of 115.788 usec -> 2022-06-24T12:13:02.403.604.512 (XO: 23.403607065)
   Max dTM/dt [1/us]:2.40e+03 at an offset of 118.438 usec -> 2022-06-24T12:13:02.403.607.162 (XO: 23.403609715)
   Min dTM/dt [1/us]:-2.31e+03 at an offset of 115.888 usec -> 2022-06-24T12:13:02.403.604.612 (XO: 23.403607165)
