-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 15:44:45 2023 UTC
-- File 20220624_115220_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.805066 to 16.325138
-each separately from 14s to 16s and from 324849us to 805355us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.519494 in 1 packets: 18
MIU packet time from 2022-06-24T11:52:37.805.066.000 to 2022-06-24T11:52:39.325.138.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.519494 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 16.1227455 to 18.1227382
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.9999927 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      C0 -> 35.6 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220624_115220_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.9278117 to 16.9281007
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.5194870  -- OK for SL --
   MIU time from 2022-06-24T11:52:37.805.066.000 to 2022-06-24T11:52:37.805.355.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.519494

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.4475877 to 18.4478767
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-24T11:52:39.324.849.000 to 2022-06-24T11:52:39.325.138.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.927811700 corrected by 40bit 10MHz counter of 16.122745500 obtained at 1Hz counter of 14. Result: 14.805069098
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-06-24T11:52:37.804.925.725
   Max 1st sample MIU time among all SL packets: 2022-06-24T11:52:39.324.997.725
   Min 1st sample XO time among all SL packets: 14.804928823
   Max 1st sample XO time among all SL packets: 16.324999395
   TM values from -266 to 200 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-24T11:52:37.804.925.725 (XO: 14.804928823 - 14.805234824, t=37.2 degC)  Rank: 00083
   TM values: Max =   200  Min =  -251  Ave =    -4.45  Std =    30.64
   Min TM value:-251 at an offset of 224.876 usec -> 2022-06-24T11:52:37.805.150.600 (XO: 14.805153699)
   Max TM value:200 at an offset of 225.138 usec -> 2022-06-24T11:52:37.805.150.863 (XO: 14.805153961)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 224.988 usec -> 2022-06-24T11:52:37.805.150.713 (XO: 14.805153811)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 120.000 usec -> 2022-06-24T11:52:37.805.045.725 (XO: 14.805048823)

 - Block number: 2 starting at 2022-06-24T11:52:39.324.708.725 (XO: 16.324710394 - 16.325016395, t=35.6 degC)  Rank: 00079
   TM values: Max =   199  Min =  -266  Ave =    -4.44  Std =    18.31
   Min TM value:-266 at an offset of 112.988 usec -> 2022-06-24T11:52:39.324.821.712 (XO: 16.324823382)
   Max TM value:199 at an offset of 110.425 usec -> 2022-06-24T11:52:39.324.819.150 (XO: 16.324820819)
   Max dTM/dt [1/us]:2.27e+03 at an offset of 113.088 usec -> 2022-06-24T11:52:39.324.821.812 (XO: 16.324823482)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 110.525 usec -> 2022-06-24T11:52:39.324.819.250 (XO: 16.324820919)
