-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 15:41:37 2023 UTC
-- File 20220624_114851_1229.dat, 695  packets found.
APID:
      1229 (EW) in 695 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 695 packets (all packets)
Total number of packets: 
      36720 in 695 packets (all packets)
Packet No from 1 to 695
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 23.139101 to 23.139101
-each separately from 23s to 23s and from 139101us to 139101us.
   Differences in neigboring packets: 
      0.000000 in 694 packets (all packets)
MIU packet time from 2022-06-24T11:49:17.139.101.000 to 2022-06-24T11:49:17.139.101.000
   Differences in neigboring packets: 
      0.000000 in 694 packets (all packets)
MIU orbit number: 
      60000 in 695 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 695 packets (all packets)
MIU orbit information / number of events: 
      0 in 695 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 695 packets (all packets)
Instrument anomalies: 
      0 (OK) in 695 packets (all packets)
FPGA configuration version: 
      211 in 695 packets (all packets)
10 MHz counter at the last 1Hz pulse from 25.1133484 to 25.1133484
   Differences in neigboring packets: 
      0.0000000 in 694 packets (all packets)
Temperature FPGA: 
      05 -> 48.5 degC in 695 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 695 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 695 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 695 packets (all packets)
Number of external events: 
      3 in 695 packets (all packets)
Number of generated alerts: 
      3 in 695 packets (all packets)
Survey memory error: 
      0 in 695 packets (all packets)
Event memory error: 
      0 in 695 packets (all packets)
Configuration error: 
      0 in 695 packets (all packets)
SDRAM CRC error: 
      0 in 695 packets (all packets)
Radio gain: 
      0 in 695 packets (all packets)
Radio bandwidth: 
      0 in 695 packets (all packets)
Digital gain: 
      0 in 695 packets (all packets)
Analog gain: 
      0 in 695 packets (all packets)

EVENT DATA READER
-- File 20220624_114851_1229.dat, 3 EW pages found. Extra packets: 83.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 3 to 3
      3 in 612 packets (all packets)
   LS 32 bits of C10MHz from descriptor (period 430s)  from 25.1969624 to 25.2073494
      Difference between the neighboring packets: 
      0.0000170 in 611 packets (all packets)
   Memory page number from 12 to 14
      Difference between the neighboring packets: 
      0 in 609 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 2 packets: 204, 408
   Event page number from 1 to 3
      Difference between the neighboring packets: 
      0 in 609 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 2 packets: 204, 408
   MIU event flag 
      1 in 612 packets (all packets)
   Time triggered event flag 
      0 in 612 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2022-06-24T11:49:17.083.612.620
   Min EW header abs. time found among all events            : 2022-06-24T11:49:17.139.101.000
   Max EW event (1st sample) abs. time found among all events: 2022-06-24T11:49:17.083.612.620
   Max EW header abs. time found among all events            : 2022-06-24T11:49:17.139.101.000
   TM values from -210 to 178 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 2 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 3 total number of packets: 612
    Page no. 1:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 3  ordered packets: 612
   TM values from -210 to 178 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=23 C1MHz=0.139101
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 611 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 25.1969624
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-06-24T11:49:17.139.101.000
   Event 1st sample abs. time : 2022-06-24T11:49:17.083.612.620)
   --- Rel. time correction for 1 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 23, 32bit 10MHz counter = 25.196962400 corrected by 40bit 10MHz counter of 15.113384900 obtained at 1Hz counter of 13. Result: 23.083614809
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              23.083614634
   Event data descriptor XO time (reconstructed seconds from XP start)    :              23.083614809
   Sampling frequency correction factor for this event:     0.999996300
--
Extremes in separate events:

 - Event number: 1 --EMPTY-- 

 - Event number: 2 --EMPTY-- 

 - Event number: 3 from 2022-06-24T11:49:17.083.612.620 to 2022-06-24T11:49:17.094.016.658 (XO: 23.083614634 - 23.094018673)
   Min TM value:-210 at an offset of 5189.319 usec -> 2022-06-24T11:49:17.088.801.939 (XO: 23.088803954)
   Max TM value:178 at an offset of 5216.457 usec -> 2022-06-24T11:49:17.088.829.076 (XO: 23.088831091)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 5189.432 usec -> 2022-06-24T11:49:17.088.802.051 (XO: 23.088804066)
   Min dTM/dt [1/us]:-1.61e+03 at an offset of 5216.569 usec -> 2022-06-24T11:49:17.088.829.189 (XO: 23.088831204)
   Med: -4.00000   Ave: -4.4732927   Std: 6.9605926
