-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 14:50:53 2023 UTC
-- File 20220624_110625_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.074174 to 20.055429
-each separately from 10s to 20s and from 55140us to 74463us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.980677 in 1 packets: 18
MIU packet time from 2022-06-24T11:06:38.074.174.000 to 2022-06-24T11:06:48.055.429.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.980677 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.7161584 to 21.7161211
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999627 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220624_110625_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.7903337 to 11.7906227
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.9806390  -- OK for SL --
   MIU time from 2022-06-24T11:06:38.074.174.000 to 2022-06-24T11:06:38.074.463.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.980677

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.7712617 to 21.7715507
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-24T11:06:48.055.140.000 to 2022-06-24T11:06:48.055.429.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 11.790333700 corrected by 40bit 10MHz counter of 11.716158400 obtained at 1Hz counter of 10. Result: 10.074175582
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996200 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-24T11:06:38.074.033.725
   Max 1st sample MIU time among all SL packets: 2022-06-24T11:06:48.055.288.725
   Min 1st sample XO time among all SL packets: 10.074035306
   Max 1st sample XO time among all SL packets: 20.055289530
   TM values from -272 to 224 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-24T11:06:38.074.033.725 (XO: 10.074035306 - 10.074341307, t=37.2 degC)  Rank: 00084
   TM values: Max =   224  Min =  -272  Ave =    -4.41  Std =    16.65
   Min TM value:-272 at an offset of 115.538 usec -> 2022-06-24T11:06:38.074.149.262 (XO: 10.074150844)
   Max TM value:224 at an offset of 113.050 usec -> 2022-06-24T11:06:38.074.146.775 (XO: 10.074148357)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 115.663 usec -> 2022-06-24T11:06:38.074.149.387 (XO: 10.074150969)
   Min dTM/dt [1/us]:-2.44e+03 at an offset of 113.163 usec -> 2022-06-24T11:06:38.074.146.887 (XO: 10.074148469)

 - Block number: 2 starting at 2022-06-24T11:06:48.054.999.725 (XO: 20.055000529 - 20.055306530, t=37.2 degC)  Rank: 00079
   TM values: Max =   198  Min =  -244  Ave =    -4.35  Std =    31.08
   Min TM value:-244 at an offset of 224.576 usec -> 2022-06-24T11:06:48.055.224.300 (XO: 20.055225104)
   Max TM value:198 at an offset of 224.288 usec -> 2022-06-24T11:06:48.055.224.013 (XO: 20.055224817)
   Max dTM/dt [1/us]:1.65e+03 at an offset of 222.213 usec -> 2022-06-24T11:06:48.055.221.938 (XO: 20.055222742)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 238.038 usec -> 2022-06-24T11:06:48.055.237.763 (XO: 20.055238567)
