-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 14:44:55 2023 UTC
-- File 20220624_110055_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.869200 to 19.317523
-each separately from 14s to 19s and from 317234us to 869489us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.447745 in 1 packets: 18
MIU packet time from 2022-06-24T11:01:12.869.200.000 to 2022-06-24T11:01:17.317.523.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.447745 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 16.0130779 to 21.0130592
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.9999813 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220624_110055_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.8822779 to 16.8825669
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.4477270  -- OK for SL --
   MIU time from 2022-06-24T11:01:12.869.200.000 to 2022-06-24T11:01:12.869.489.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.447745

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.3302939 to 21.3305829
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-24T11:01:17.317.234.000 to 2022-06-24T11:01:17.317.523.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.882277900 corrected by 40bit 10MHz counter of 16.013077900 obtained at 1Hz counter of 14. Result: 14.869203216
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-24T11:01:12.869.059.725
   Max 1st sample MIU time among all SL packets: 2022-06-24T11:01:17.317.382.725
   Min 1st sample XO time among all SL packets: 14.869062941
   Max 1st sample XO time among all SL packets: 19.317384599
   TM values from -268 to 186 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-24T11:01:12.869.059.725 (XO: 14.869062941 - 14.869368942, t=37.2 degC)  Rank: 00081
   TM values: Max =   182  Min =  -222  Ave =    -4.35  Std =    29.53
   Min TM value:-222 at an offset of 198.951 usec -> 2022-06-24T11:01:12.869.258.675 (XO: 14.869261891)
   Max TM value:182 at an offset of 198.676 usec -> 2022-06-24T11:01:12.869.258.400 (XO: 14.869261616)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 157.713 usec -> 2022-06-24T11:01:12.869.217.438 (XO: 14.869220654)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 142.663 usec -> 2022-06-24T11:01:12.869.202.388 (XO: 14.869205604)

 - Block number: 2 starting at 2022-06-24T11:01:17.317.093.725 (XO: 19.317095598 - 19.317401599, t=37.2 degC)  Rank: 00088
   TM values: Max =   186  Min =  -268  Ave =    -4.39  Std =    20.44
   Min TM value:-268 at an offset of 115.125 usec -> 2022-06-24T11:01:17.317.208.850 (XO: 19.317210724)
   Max TM value:186 at an offset of 112.613 usec -> 2022-06-24T11:01:17.317.206.337 (XO: 19.317208211)
   Max dTM/dt [1/us]:2.36e+03 at an offset of 115.250 usec -> 2022-06-24T11:01:17.317.208.975 (XO: 19.317210849)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 112.738 usec -> 2022-06-24T11:01:17.317.206.462 (XO: 19.317208336)
