-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 14:20:04 2023 UTC
-- File 20220624_104710_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.525609 to 17.545648
-each separately from 7s to 17s and from 525609us to 545648us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.019461 in 1 packets: 18
MIU packet time from 2022-06-24T10:47:20.525.609.000 to 2022-06-24T10:47:30.545.648.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.019461 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.7622118 to 18.7621743
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999625 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220624_104710_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.2878218 to 9.2881108
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.0194230  -- OK for SL --
   MIU time from 2022-06-24T10:47:20.525.609.000 to 2022-06-24T10:47:20.525.898.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.019461

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.3075338 to 19.3078228
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-24T10:47:30.545.359.000 to 2022-06-24T10:47:30.545.648.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.287821800 corrected by 40bit 10MHz counter of 8.762211800 obtained at 1Hz counter of 7. Result: 7.525611945
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996200 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-24T10:47:20.525.468.725
   Max 1st sample MIU time among all SL packets: 2022-06-24T10:47:30.545.507.725
   Min 1st sample XO time among all SL packets: 7.525471669
   Max 1st sample XO time among all SL packets: 17.545510298
   TM values from -256 to 226 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-24T10:47:20.525.468.725 (XO: 7.525471669 - 7.525777670, t=37.2 degC)  Rank: 00080
   TM values: Max =   174  Min =  -256  Ave =    -4.47  Std =    18.30
   Min TM value:-256 at an offset of 120.188 usec -> 2022-06-24T10:47:20.525.588.912 (XO: 7.525591857)
   Max TM value:174 at an offset of 117.575 usec -> 2022-06-24T10:47:20.525.586.300 (XO: 7.525589245)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 120.275 usec -> 2022-06-24T10:47:20.525.589.000 (XO: 7.525591945)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 117.713 usec -> 2022-06-24T10:47:20.525.586.437 (XO: 7.525589382)

 - Block number: 2 starting at 2022-06-24T10:47:30.545.218.725 (XO: 17.545221297 - 17.545527298, t=37.2 degC)  Rank: 00081
   TM values: Max =   226  Min =  -251  Ave =    -4.41  Std =    20.52
   Min TM value:-251 at an offset of 120.875 usec -> 2022-06-24T10:47:30.545.339.600 (XO: 17.545342172)
   Max TM value:226 at an offset of 118.375 usec -> 2022-06-24T10:47:30.545.337.100 (XO: 17.545339672)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 121.000 usec -> 2022-06-24T10:47:30.545.339.725 (XO: 17.545342297)
   Min dTM/dt [1/us]:-2.43e+03 at an offset of 118.488 usec -> 2022-06-24T10:47:30.545.337.212 (XO: 17.545339785)
